Braun Thermoscan Type 6013
Here you can find all about Braun Thermoscan Type 6013 like manual and other informations. For example: .
Braun Thermoscan Type 6013 manual (user guide) is ready to download for free.
On the bottom of page users can write a review. If you own a Braun Thermoscan Type 6013 please write about it to help other people. [ Report abuse or wrong photo | Share your Braun Thermoscan Type 6013 photo ]
Braun Thermoscan Type 6013, size: 290 KB
Braun Thermoscan Type 6013
User reviews and opinions
|alegomes||9:53pm on Tuesday, September 28th, 2010|
|the phone is very user friendly,faster processor after upgrading to froyo2.2 software. I am about to trade in my problematic Nokia N8 only one month old. So now got to choose which: Galaxy 9000 or this HTC?|
|curly_joe||12:43pm on Thursday, August 19th, 2010|
|Bad quality of the device and service in the selling office. External design is fine Not reliable mechanical/Electrinic of the device.|
|Chabbrik||3:14am on Thursday, July 29th, 2010|
|milsy, then you need to come out to where I live. So far only phone that gets any reception is Nokia 6120! Since I bought my Desire I have been crazy in love with it!|
|mir||3:01am on Tuesday, July 20th, 2010|
|HTC Desire After having to wait for 4 weeks in Sweden I decided to purchase from Amazon (my 1st time) and was happily surprised. GREAT! I recieved the item a couple of days before I expected it so YAY! and the phone was in excellent condition. It had everything it needed e.|
|newpgr||5:37pm on Wednesday, July 14th, 2010|
|The only way I have resolved this, and it has happened at least three times, is to pull of the back cover and pop the battery. And, for the record.|
|agrieco||3:51am on Friday, July 9th, 2010|
|Great Phone with a miner cons, can be fix in the future. touch sensitivity. it is a good mobile but,,, where is the secondary camera for a smartphone???? browser, touch sensor SECONDARY camera|
|Jairo Pereira||2:08pm on Sunday, June 6th, 2010|
|Love this phone Fast, Responsive, great screen, amazing camera in video or pics No slide out keyboard This is not a OLED version this is a SLCD version! very good no - so far|
|cobra_sa||10:07pm on Saturday, May 22nd, 2010|
|"I have it for 3 weeks now and its the best thing ive ever held in my hands.My friend has iPhone 4,and its a great phone but Desire is much better. "I love my desire! some of the software updates have slowed it down considerably tho, anyone have the same issue? It feels lovely in my hand.|
|LarkinVB||10:07pm on Monday, May 17th, 2010|
|Its and amaz phone...buy it widout thinking. Fast, Good touch Screen, looks, syle Battery time is very low....lasts a day Does it all quickly Battery life is not so good, and the carrier could be better|
Comments posted on www.ps2netdrivers.net are solely the views and opinions of the people posting them and do not necessarily reflect the views or opinions of us.
IRT 3020 IRT 3520
Type 6012 Type 6013
Internet: www.braun.com Manufactured by: Braun GmbH Frankfurter Str. Kronberg / Germany
6-012-136/00/X-05 USA Printed in Germany
oS ca n
Th erm oS c an
The Braun ThermoScan thermometer has been carefully developed for accurate, safe and fast human body temperature measurements in the ear. The shape of the thermometer prevents it from being inserted too far into the ear canal to damage the eardrum. However, as with any thermometer, proper technique is critical to obtaining accurate temperatures. Therefore, please read all instructions carefully and thoroughly before using this product and keep the manual handy for future reference.
The operating ambient temperature range for this thermometer is 50 F to 104 F (10 C to 40 C). Do not expose the thermometer to temperature extremes below - 4 F or over 122 F ( - 20 / 50 C) or excessive humidity ( > 95 % RH non-condensing). This thermometer must only be used with genuine Braun ThermoScan Lens Filters (LF 40). Never use this thermometer without a new, clean lens filter attached. If the thermometer is ever accidentally used without a lens filter attached, clean the lens (see Care and cleaning). Always store thermometer in travel case, protective cover or plastic case. Basic safety precautions should always be observed, especially when using the thermometer on or near children and disabled persons. Keep lens filters out of reach of children. This thermometer is intended for home use only. Use of this thermometer is not intended as a substitute for consultation with your physician. Do not leave thermometer or lens filters with infants or children at any time.
How does Braun ThermoScan work?
The Braun ThermoScan thermometer measures the infrared heat generated by the eardrum and surrounding tissue. To help ensure accuracy, the thermometer scans by taking 8 measurements in just one second and displaying the highest temperature. The displayed ThermoScan temperature is the actual measured ear canal temperature plus a mathematical adjustment to approximate the familiar oral range. However, this is not necessarily the same as an oral temperature measured at the same time.
Why measure in the ear?
The goal of thermometry is to measure core body temperature, which is the temperature of the vital organs. Clinical studies have shown that the ear is an excellent site to measure body temperature, since ear temperatures reflect core body temperature. The eardrum shares blood supply with the temperature control center in the brain, the hypothalamus, allowing changes in body temperature to be reflected sooner in the ear than at other sites. 4
Axillary temperatures reflect skin temperature which may not indicate core body temperature. Oral temperatures are influenced by drinking, eating and breathing through the mouth. Rectal temperatures often lag behind changes in core body temperature and there is a risk of cross-contamination.
Normal body temperature is a range. The following table1 shows that ranges of normal also vary by site. Therefore, readings from different sites, even if taken at the same time, should not be directly compared. Axillary: Oral: Rectal: ThermoScan2: 94.5 F - 99.1 F 95.9 F - 99.5 F 97.9 F - 100.4 F 96.4 F - 100.4 F 34.7 C - 37.3 C 35.5 C - 37.5 C 36.6 C - 38.0 C 35.8 C - 38.0 C
Also, a persons normal temperature range tends to decrease with age. The following table shows normal ThermoScan ranges by age. Normal ThermoScan temperature ranges- 2 years 3 - 10 years 11 - 65 years > 65 years 97.5 F - 100.4 F 97.0 F - 100.0 F 96.6 F - 99.7 F 96.4 F - 99.5 F 36.4C - 38.0 C 36.1C - 37.8 C 35.9C - 37.6 C 35.8C - 37.5 C
However, the range of normal also varies from person to person and fluctuates throughout the day. It is therefore important to determine your and your family members normal temperature ranges. This is easily done using Braun ThermoScan. Practice taking temperatures on yourself and healthy family members to determine their normal temperature ranges. Note: When consulting your physician, communicate that the ThermoScan temperature is a temperature measured in the ear and if possible, note the individuals normal ThermoScan temperature range as additional reference.
1. Chamberlain, J.M., Terndrup, T.E., New Light on Ear Thermometer Readings, Contemporary Pediatrics, March 1994. 2. Chamberlain, J.M., et al., Determination of Normal Ear Temperature with an Infrared Emission Detection Thermometer, Annals of Emergency Medicine, January 1995.
1. 2. 3. 4. 5. 6. 7. 8. 9. 10. 11. 12. 13. Lens filter Probe tip Probe Lens filter detector Lens filter ejector Display (LCD) LCD light button (IRT 3520 only) /mem button (On/memory function) Battery door lock Battery door Activation button Label Travel case/protective cover/storage pouch (depending on configuration)
Braun ThermoScan thermometer Use and Care manual Quick Reference Guide Lens filters (20, plus one on the thermometer) Owner registration/warranty card Additional items may be included as noted on outer packaging Label The first time you use the thermometer, please make sure to apply the special label included in the package, in the language of your choice (Item 12 on page 3).
How to use Braun ThermoScan
1. Always make sure a new, clean lens filter is in place to help ensure an accurate reading. The thermometer will not function without a lens filter attached (see Changing the lens filter). 2. Press the /mem button. The LCD (liquid crystal display) is activated, showing all segments. When the ready symbol is displayed the thermometer is ready for use.
To help ensure an accurate temperature reading, use the following technique: If you are right handed, hold the thermometer in the right hand and take the temperature in the right ear. If you are left handed, hold it in the left hand and use the left ear. 3. Perform an ear tug to straighten the ear canal. This gives the thermometer a clear view of the eardrum.
Children under 1 year: Pull the ear straight back. Children aged 1 year to adult: Pull the ear up and back. An ear tug is best performed by using your free hand to grasp the outer edge of the top half of the ear. To take your own temperature, wrap your free hand around the back of your head and grab your ear from behind. 4. While tugging the ear, fit the probe snugly into the ear canal as far as possible and press the activation button. Release it when you hear a beep. This is the Temp Beep that confirms the end of measurement. 5. Remove the thermometer from the ear canal. The LCD displays the temperature measured.
6. Replace the lens filter after each measurement: press the ejector button and put on a new, clean lens filter.
7. A new measurement can be taken as long as the ready symbol is shown. If no button is pressed within 2 minutes, the thermometer will turn off automatically. 7
Note: In the following situations, it is recommended that you take three temperatures in the same ear. If they differ, use the highest reading. Infants in the first 90 days of life. Children under three years of age who have a condition such as a compromised immune system and for whom the presence or absence of fever is critical. When you are first learning to use the ear thermometer until you are comfortable with the technique and are obtaining consistent readings. Important: As with any type of thermometer, slight temperature variations ( 0.3 - 0.5 F / 0.2 - 0.3 C) can occur, if consecutive measurements are taken.
Temperature taking hints
The right ear reading may differ from the reading taken at the left ear. Therefore, always take the temperature in the same ear. The ear must be free from obstructions or excess earwax buildup to take an accurate reading. External factors may influence ear temperatures, including when an individual has: - been lying on one ear or the other - had their ears covered - been exposed to very hot or very cold temperatures, or - been recently swimming or bathing. In these cases, remove the individual from the situation and wait 20 minutes prior to taking a temperature.
The last temperature taken before the thermometer powers down is stored in memory. To enter the memory mode, press the /mem button. Even in memory mode, a new temperature can be taken provided that the ready symbol is shown. IRT 3020 The last stored temperature is displayed along with the MEM symbol. To quit the memory mode, press the /mem button again. IRT 3520 This model allows you to store up to 8 temperatures. When pressing the /mem button, the display shows the memory cell number (e.g. MEM 1). When releasing the /mem button, the stored temperature is displayed. Each time the /mem button is pressed, a new memory cell is displayed (up to MEM 8).
An empty memory cell shows - - - F. Only the first empty memory cell will be displayed. To quit the memory mode, press the /mem button again after reaching MEM 8 or - - - F. Memory clear Press the /mem button for 5 seconds to clear the temperatures stored in memory. Release the /mem button to return to the ready symbol. LCD light (model IRT 3520 only) For easy nighttime reading.
Changing the temperature scale
This thermometer is shipped with the Fahrenheit temperature scale activated. If you wish to switch to Celsius (C), proceed as follows: Turn on the thermometer. (If it is already turned on, make sure it is not in memory mode.) Press the /mem button and keep it pressed. Then press and release the activation button to switch over to the change temperature scale mode. By pressing the activation button again, the Celsius scale is activated, C is displayed on the LCD. Each further pressing of the activation button toggles the scale between C and F. IRT 3020 Release the /mem button to return to the ready symbol.
All temperatures stored in memory will automatically be converted to the selected scale when displayed on the LCD. IRT 3520
Changing the lens filter
The thermometer is supplied with a lens filter in place. To assure accuracy and sanitary practice, Braun recommends replacing the lens filter after each use. To install a new lens filter, first remove the one in place by pressing the lens filter ejector. Check the lens for any damage. Then take a new lens filter from the box, and holding it by its edges, slide it onto the probe until it snaps in place. Caution: Do not touch the tip of the probe or lens filters when installing. Fingerprints, ear wax, dust and other soiling compounds reduce the transparency, resulting in lower temperature readings (see also Care and cleaning). Should you run out of lens filters and need to take a temperature, you may use the following lens filter cleaning procedure: Clean the lens filter without removing it from the thermometer with a soft cloth or cotton swab moistened with alcohol. Do not use hot or boiling water. Dry completely with a soft cloth before reusing. Replace lens filter as soon as possible with a new one (LF 40 ). Additional lens filters (LF 40 ) are available at most stores 10
carrying Braun ThermoScan thermometers or at Braun Service Centers. If you have accidentally placed the thermometer in your ear without a lens filter in place, be sure and clean the probe (see Care and cleaning and then apply a clean lens filter.
Care and cleaning
The probe tip is the most delicate part of the thermometer. It has to be clean and intact to ensure accurate readings. If the thermometer is ever accidentally used without a lens filter, clean the lens as follows: Hold the appliance with the probe tip facing down to prevent liquid from entering the probe tip area. Very gently wipe the surface with a cotton swab or soft cloth moistened with alcohol. After cleaning, allow at least 45 minutes drying time before reattaching a new, clean lens filter and taking a temperature. If the probe tip is damaged, contact Braun. Use a soft, dry cloth to clean the thermometer display and exterior. Do not use abrasive cleaners. Never submerge the thermometer in water or any other liquid. Store thermometer and lens filters in a dry location free from dust and contamination and away from direct sunlight. The ambient temperature at the storage location should remain fairly constant and within the range of 50 F to 104 F (10 C - 40 C). Always keep cleaning solutions and rubbing alcohol away from children.
Replacing the batteries
The thermometer is supplied with two 3 V lithium batteries (CR 2032/DL 2032). Insert new batteries when the battery symbol appears on the display. Using the tip of a ball-point pen, press the battery door lock to open the battery compartment. Remove the batteries and replace with new batteries, making sure the poles are in the right direction. Slide battery door back in until it snaps in place.
Error message Situation No lens filter is attached. Solution Attach new, clean lens filter
No lens filter is attached and activation button was pressed while probe was in the ear.
Make sure probe tip is clean; refer to section Care and cleaning if necessary. Attach new, clean lens filter to stop error beeps. Allow the thermometer to remain in a room for 30 minutes where the temperature is between 50 F to 104 F (10 C - 40 C) Make sure new, clean lens filter is attached and thermometer is properly inserted. Then, take a new temperature.
Ambient temperature is not within the allowed operating range 50 F to 104 F (10 C - 40 C) or changing too rapidly. Temperature taken is not within typical human temperature range (93.2 F 108 F / 34 C - 42.2 C). HI = too high LO = too low
Battery is low, but Insert new batteries thermometer will still operate correctly. (No light function on IRT 3020 models.) Battery is too low to take correct temperature readings. System error If error persists If error still persists Insert new batteries
Wait 2 minutes until thermometer powers down, then turn on again. reset the thermometer by removing the batteries and putting them back in. call Braun ThermoScan Customer Service at 1- 800 - 327-7226.
The Braun ThermoScan thermometer has been shown in clinical studies to obtain accurate temperatures on persons of all ages. However, there are certain situations when the ear thermometer should not be used. These include but may not be limited to the following situations. If there is blood or drainage in the external ear canal. For persons who have deformities of the face and ear canal where the probe of the thermometer cannot be inserted fully into the ear canal. For persons wearing hearing aids or ear plugs, remove the device and wait 20 minutes prior to taking a temperature. Use the untreated ear if prescription ear drops or other ear medications have been placed in the ear canal. Never attempt to clean inside ears. You could damage the ear drum or surrounding tissues. You should remove excess ear wax only when you can reach it with a wash cloth. If you suspect that you or your child has excess ear wax, consult your physician.
Many persons may not have an elevated temperature even when they are ill. These include, but are not limited to infants under 90 days of age, persons taking steroids, antibiotics or antipyretics (acetaminophen, ibuprofen, aspirin), persons with compromised immune systems, including the elderly or persons with some chronic illnesses. Consult your physician if you feel an illness is present even if there may not be an elevated temperature. Fever is described as an elevation of body temperature over an individuals normal temperature. An elevated temperature or fever is often viewed as a danger sign. In fact, a fever can be very beneficial, and helps our immune system work more effectively. A fever should be evaluated in the light of other physical symptoms. With the exception of newborn infants, the presence of fever, without any other symptoms of illness, or in a child who is behaving normally may not be cause for concern. On the other hand, a physician should be consulted in the following situations: - vomiting - diarrhea - changes in appetite, activity or breathing, or - with children who are irritable, lethargic or unusually sleepy. 13
Some people, like the elderly, infants under 90 days of age, those with compromised immune systems, or persons who take steroids, for example are often unable to build a response to illness or environmental conditions. These individuals may not have a fever when they are ill, or their elevated temperature may be lower than expected for the severity of their illness. Other medications such as anti-inflammatories and some analgesics may also mask fever. The presence or absence of fever should rarely be used as the only measurement of the significance of illness. Your physician should be contacted whenever there is a question about your familys health. Antipyretics, like acetaminophen or ibuprofen, are usually recommended to relieve the aches and associated symptoms of fever, not the fever itself. Febrile seizures, or convulsions, which usually occur in children six months to six years of age, are thought to occur not because a fever is present, but because of the rate of rise of the childs temperature. Call your physician if your child has a febrile seizure or you desire further information. Use of the Braun ThermoScan thermometer is not intended as a substitute for consultation with your physician.
Displayed temperature range Operating ambient temperature range Display resolution 93.2 F -108 F (34 C - 42.2 C) 50 F -104 F 0.1 F or C (10 C - 40 C)
Accuracy for patient temperature range Maximum Laboratory Error 96.8 F -102.2 F (36 C - 39 C): 0.4 F 0.2 C outside this range: 0.5 F 0.3 C Long term storage ranges Temperature: Humidity: Battery life: - 4 F to 122 F ( - 20 C to 50 C) 95 % non-condensing 2 years / 1000 measurements
This infrared thermometer meets requirements established in ASTM Standard E 1965-98 (for the thermometer system [thermometer with lens filter]). Full responsibility for the conformance of the product to the standard is assumed by Braun GmbH, 61476 Kronberg, Germany. ASTM laboratory accuracy requirements for the thermometer only (not including a lens filter) in the display range of 98.8 F to 102.2 F (36 C to 39 C) for infrared thermometers is 0.4 F ( 0.2 C), whereas for mercury-in-glass and electronic thermometers, the requirement per ASTM Standards E 667-86 and E 1112-86 is 0.2 F ( 0.1 C). This appliance conforms to the following standards: DIN EN 60601-1: 3/96 Medical electrical equipment Part 1: General requirements for safety DIN EN 12470-5: 2003 Clinical thermometers Part 5: Performance of infrared ear thermometers (with maximum device)
TYPE BF EQUIPMENT
Attention, consult MEDICAL ELECTRICAL EQUIPMENT ACCOMPANYING CLASSIFIED BY UNDERWRITERS LABORATORIES INC. DOCUMENTS WITH RESPECT TO ELECTRIC SHOCK, FIRE AND MECHANICAL HAZARDS ONLY IN ACCORDANCE WITH UL 2601-1 / CAN CSA C22.2 No.601.1 10HK
Internally Powered Equipment Continuous Operation Not Protected against Ingress of Water U.S. Patent No. 5,088,834 Other Patents Pending MEDICAL ELECTRICAL EQUIPMENT needs special precautions regarding EMC. For detailed description of EMC requirements please contact the Braun ThermoScan Customer Service at 1-800-327-7226. Portable and mobile RF communications equipment can affect MEDICAL ELECTRICAL EQUIPMENT. In the unlikely event you experience any difficulty using the Braun ThermoScan thermometer, simply call us toll free at 1-800-327-7226.
The Braun ThermoScan thermometer has a limited three year warranty. Should service be required during or after the warranty period, you must call Braun ThermoScans Customer Service Department at 1-800-327-7226 to obtain a return authorization number and shipping address. Repackage the thermometer carefully in its original packaging or securely packed to avoid damage during shipping. Include the original sales slip indicating the date of purchase, a note describing the problem, your authorization number, and your return address. Send the Braun ThermoScan thermometer prepaid and insured. Note: Shipping address varies by country. Answers to any question regarding operation of the Braun ThermoScan thermometer may be obtained by calling us toll free at: 1-800-327-7226 Monday-Friday 6:30 AM to 5:00 PM Pacific Time
Limited three year warranty
Braun warrants this product against any defects that are due to faulty material or workmanship for a period of three years from the original date of consumer purchase or receipt as a gift. This warranty applies when used for normal household use in accordance with the Use and Care Manual and excludes the battery and damage to the product resulting from accident or misuse. This product is not warranted when used in a professional environment. In no event shall Braun be liable for any special, incidental, indirect or consequential damages in connection with the purchase or use of this product or costs over the original cost of the product. If the product should become defective within the warranty period, contact Braun ThermoScans Customer Support Department at 1-800-327-7226 for repair or replacement at no charge. Braun reserves the right to replace a defective product with the most comparable product currently available. This product has a five year minimum expected lifetime. While this warranty gives you specific legal rights, you may also have other rights which vary from state to state/province to province.
IRT 4520/4020 EK KURTZ DESIGN 12.02.03
IRT 4520 IRT 4020
Type 6022 Type 6023
6022230_IRT_MN_S2 Seite 1 Mittwoch, 2. Februar 2005 12:32 12
Braun Infoline English Franais Espaol Portugus 4, 52 13, 52 22, 52 31, 52 51, 52 GB Helpline Should you require any further assistance please call Braun (UK) Consumer Relations on IRL Helpline Should you require any further assistance please call Braun Consumer Relations on F Vous avez des questions sur ce produit ? Appelez Braun France 70 Vous avez des questions sur ce produit ? Appelez Braun Belgique (02) 11 Servicio al consumidor para Espaa: 9 01-84 Mexico: 00 Argentina: 44 Gillette (3) Internet: www.braun.com P Manufactured by: Braun GmbH Frankfurter Str. 145 D-61476 Kronberg / Germany Chile: Servio ao Consumidor para Portugal: 00 33
6-022-230/00/I-05/M GB/F/E/P/Arab Printed in Germany
n ca oS m er Th
6022230_IRT_MN_S4-66 Seite 4 Mittwoch, 2. Februar 2005 12:37 12
The Braun ThermoScan thermometer has been carefully developed for accurate, safe and fast temperature measurements in the ear. The shape of the thermometer prevents it from being inserted too far into the ear canal to hurt the eardrum. However, as with any thermometer, proper technique is critical to obtaining accurate temperatures. Therefore, read the use instructions carefully and thoroughly. Important The operating ambient temperature range for this thermometer is 1040 C (50104 F). Do not expose the thermometer to temperature extremes (below 20 C / 4 F or over 50 C / 122 F) nor excessive humidity (> 95 % RH). This thermometer must only be used with genuine Braun ThermoScan Lens Filters (LF 40). Never use this thermometer without a new, clean lens filter attached. Keep lens filters out of reach of children. This thermometer is intended for household use only. Use of this thermometer is not intended as a substitute for consultation with your physician.
How does Braun ThermoScan work?
Braun ThermoScan measures the infrared heat generated by the eardrum and surrounding tissues. To avoid gross temperature differences, the sensor itself is brought to a temperature close to that of the human body. When the Braun ThermoScan is placed in the ear, it continuously monitors the infrared radiation. The measurement will be finished and the result is displayed, when an accurate reading can be assured.
Why measure in the ear?
The goal of thermometry is to measure core body temperature which is the temperature of the vital organs. Ear temperatures accurately reflect core body temperature, since the eardrum shares blood supply with the temperature control center in the brain, the hypothalamus. Therefore, changes in body temperature are reflected sooner and more accurately in the ear than at other sites. Axillary temperatures only measure skin temperature and therefore, are not a reliable indicator of core body temperature. Oral temperatures are influenced by drinking, eating and breathing. Rectal temperatures lag behind changes in core body temperature and there is a risk of cross-contamination.
6022230_IRT_MN_S4-66 Seite 5 Mittwoch, 2. Februar 2005 12:37 12
Normal body temperature is a range. The following table shows that this normal range also varies by site. Therefore, readings from different sites should not be directly compared. Normal ranges by site: Axillary: Oral: Rectal: ThermoScan: 34.7 37.3 C 35.5 37.5 C 36.6 38.0 C 35.8 38.0 C 94.5 99.1 F 95.9 99.5 F 97.9 100.4 F 96.4 100.4 F
As well, a persons normal temperature range tends to decrease with age. The following table shows normal ThermoScan ranges by age. Normal ThermoScan ranges by age: years years years > 65 years 36.4 38.0 C 36.1 37.8 C 35.9 37.6 C 35.8 37.5 C 97.5 100.4 F 97.0 100.0 F 96.6 99.7 F 96.4 99.5 F
However, the range of normal also varies from person to person and fluctuates throughout the day. It is therefore important to determine your normal temperature range. This is easily done using Braun ThermoScan. Practice taking temperatures on yourself and healthy family members to determine their normal temperature ranges. Note: When consulting your physician, communicate that the ThermoScan temperature is a temperature measured in the ear and if possible, note the individuals normal ThermoScan temperature range as additional reference.
6022230_IRT_MN_S4-66 Seite 6 Mittwoch, 2. Februar 2005 12:37 12
1. 2. 3. 4. 5. 6. 7. 8. 9. 10. 11. Lens filter Probe Lens filter detector Lens filter ejector Display I/O button (On/memory function IRT 4520 only) ExacTemp light start button Battery door Protective cover (IRT 4520) Cap (IRT 4020)
How to use your Braun ThermoScan?
1. To achieve accurate readings, make sure a new, clean lens filter (1) is in place before each measurement.
2. IRT 4020: Push the start button (8). IRT 4520: Push the I/O button (6). Then wait for the ready signal beep and the ready symbol in the display.
3. Fit the probe snuggly into the ear canal, then push the start button (8). If the probe has been fitted securely into the ear canal during the complete measuring process, a long beep will signal the end of the measuring process. You can be assured to have taken an accurate temperature reading. The result is shown on the display (5).
n ca oS erm Th
6022230_IRT_MN_S4-66 Seite 7 Mittwoch, 2. Februar 2005 12:37 12
If you take the temperature of another person, the ExacTemp light (7) will be of help. It flashes during the measuring process as long as the probe is securely positioned, and lights up continuously when an accurate reading has been taken. 4. If the probe has not been constantly placed in a stable position in the ear canal, a sequence of short beeps will sound, the ExacTemp light will go out and the display will show an error message (POS = position error). 5. For the next measurement, eject the used lens filter (push ejector (4)) and put on a new, clean lens filter. IRT 4020: Clear the display by pushing the start button once. IRT 4520: Clear the display by pushing the I/O button once. Wait for the ready signal. Fit the probe snuggly into the ear canal, then push the start button. The Braun ThermoScan ear thermometer turns off automatically after 60 seconds of inactivity. The IRT 4520 can also be turned off by pressing the I/O button for at least three seconds. The display will shortly flash OFF and after releasing the button it will go blank.
Temperature taking hints
The right ear reading may differ from the reading taken at the left ear. Therefore, always take the temperature in the same ear. The ear must be free from obstructions or excess earwax build-up to take an accurate reading. External factors may influence ear temperatures, including when an individual has: been lying on one ear or the other had their ears covered been exposed to very hot or very cold temperatures, or been recently swimming or bathing. In these cases, remove the individual from the situation and wait 20 minutes prior to taking a temperature. Use the untreated ear if prescription ear drops or other ear medications have been placed in the ear canal. 7
6022230_IRT_MN_S4-66 Seite 8 Mittwoch, 2. Februar 2005 12:37 12
The last temperature taken is stored in its memory and will be automatically displayed when it is turned on again. The display will show MEM.
IRT 4520: This model stores the last 8 temperature readings. To display the stored readings, the thermometer must be turned on. Then press the I/O button for at least 1 second. The display shows the memory number (e. g. MEM 1), and when releasing the I/O button, the stored temperature for that memory number is displayed, together with MEM. If I/O button is pressed too long, the thermometer will be switched off. Each further pressing of the I/O button displays the remaining memory numbers (up to MEM 8). MEM 1 is the latest reading, MEM 8 would be the oldest. The memory mode is left automatically after displaying the oldest reading, or after pressing the I/O button for at least 1 second.
Changing the temperature scale
Your Braun ThermoScan is shipped with the Celsius temperature scale activated. If you whish to switch to Fahrenheit (F), proceed as follows: Make sure the thermometer is turned off. Press and hold down the start button (IRT 4020) / the I/O button (IRT 4520). After about 3 seconds the display will show this sequence: C / SET / F / SET. Release the start button / I/O button when F is shown. There will be a short beep to confirm the new setting, then the thermometer is turned off automatically. When turned on again, temperatures will be shown in Fahrenheit, until the above procedure is repeated.
6022230_IRT_MN_S4-66 Seite 9 Mittwoch, 2. Februar 2005 12:37 12
Care and cleaning
The probe tip is the most delicate part of the thermometer. It has to be clean and intact to ensure accurate readings. If the thermometer is ever accidentally used without a lens filter, clean the probe tip as follows: Very gently wipe the surface with a cotton swab or soft cloth moistened with alcohol. After the alcohol has completely dried out, you can put a new lens filter on and measure. If the probe tip is damaged, contact Braun. Use a soft, dry cloth to clean the thermometer display and exterior. Do not use abrasive cleaners. Never submerge this thermometer in water or any other liquid. Store thermometer and lens filters in a dry location free from dust and contamination and away from direct sunlight. Additional lens filters (LF 40) are available at most stores carrying Braun ThermoScan or at Braun Service Centers.
6022230_IRT_MN_S4-66 Seite 10 Mittwoch, 2. Februar 2005 12:37 12
Error message Situation No lens filter is attached. Solution Attach new, clean lens filter.
The thermometer cannot identify a secure position of the probe. An accurate measurement was not possible. POS = position error
IRT 4020: Clear the display by pushing the start button once. IRT 4520: Clear the display by pushing the I/O button once. Take care that the positioning of the probe is correct and remains stable. Allow the thermometer to remain in a room for 30 minutes where the temperature is between 10 and 40 C or 50 and 104 F.
Ambient temperature is not within the allowed operating range (C or F).
Temperature taken is not within typical human temperature range (34 42.2 C or 93.2108 F). HI = too high
Make sure a new, clean lens filter is attached and thermometer is properly inserted. Then, take a new temperature.
LO = too low System error Wait 1 minute until the thermometer turns off automatically, then turn on again. reset the thermometer by removing the batteries and putting them back in. have the thermometer checked at a Braun Customer Service Center.
If error persists,
If error still persists,
6022230_IRT_MN_S4-66 Seite 11 Mittwoch, 2. Februar 2005 12:37 12
Error message Situation Battery is low, but thermometer will still operate correctly. Solution Insert new batteries.
Battery is too low to take correct temperature readings.
Insert new batteries.
Do you have any further questions?
Call Braun Infoline, please see page 2.
Replacing the batteries
Ther moS can
The thermometer is supplied with two 1.5 V type AA (LR 6). For best performance, we recommend Duracell alkaline batteries. Insert new batteries when the battery symbol appears on the display.
Open the battery compartment. Remove the batteries and replace with new batteries, making sure the poles are in the right direction. Slide battery door back until it snaps in place.
To protect the environment, dispose of empty batteries at your retail store or at appropriate collection sites according to national or local regulations.
The thermometer is initially calibrated at the time of manufacture. If this thermometer is used according to the use instructions, periodic re-adjustment is not required. If at any time you question the accuracy of measurement, please contact Braun Infoline, phone (see page 2). Manufacturing date is given by the LOT number located in the battery compartment. The first number after LOT represents the last digit of the year of manufacture. The next three digits give the day of the year of manufacture. An example: LOT 0116 xx xx this product has been manufactured on the 116th day of the year 2000.
6022230_IRT_MN_S4-66 Seite 12 Mittwoch, 2. Februar 2005 12:37 12
Displayed temperature range: Operating ambient temperature range: Display resolution: Accuracy for displayed temperature range: Clinical repeatability: Battery life: 3442.2 C (93.2108 F) 1040 C (50104 F) 0.1 C or F 0.2 C (35.542 C) (95.9107.6 F) 0.3 C (outside this temperature range) 0.14 C ( 0.26 F) 2 years / 1000 measurements
Type BF equipment Subject to change without notice. This appliance conforms to the following standards: DIN EN 60601-1: 3/96 Medical electrical equipment Part 1: General requirements for safety DIN EN 12470-5: 2003 Clinical thermometers Part 5: Performance of infrared ear thermometers (with maximum device) This product conforms to the provisions of the EC directive 93/42/EEC (Medical Device Directive). MEDICAL ELECTRICAL EQUIPMENT needs special precautions regarding EMC. For detailed description of EMC requirements please contact the Braun Infoline (see page 2). Portable and mobile RF communications equipment can affect MEDICAL ELECTRICAL EQUIPMENT. At the end of the product's useful life, please dispose of it at appropriate collection points provided in your country. Guarantee We grant 2 years guarantee on the product commencing on the date of purchase. Within the guarantee period we will eliminate, free of charge, any defects in the appliance resulting from faults in materials or workmanship, either by repairing or replacing the complete appliance as we may choose. This guarantee extends to every country where this appliance is supplied by Braun or its appointed distributor. This guarantee does not cover: damage due to improper use, normal wear or use as well as defects that have a negligible effect on the value or operation of the appliance. The guarantee becomes void if repairs are undertaken by unauthorised persons and if original Braun parts are not used. To obtain service within the guarantee period, hand in or send the complete appliance with your sales receipt to an authorised Braun Customer Service Centre. For UK only: This guarantee in no way affects your rights under statutory law. 12
6022230_IRT_MN_S4-66 Seite 52 Mittwoch, 2. Februar 2005 12:37 12
Guarantee and Service Centers Bureaux de garantie et centrales service aprs-vente Ocinas de garantia y ocinas centrales del servicio Entidade de garantia e centros de servio
Korea Gillette Korea Ltd. 144-27 Samsung-dong, Kangnam-ku, Seoul, Korea, 080-920-6000 Kuwait Union Trading Company, Braun Service Center, P. O. Box 28 Safat, Safat Code 13001, Kuwait, Latvia Latintertehservice Co., 72 Bullu Street, House 2, Riga 1067, Lebanon Magnet SAL - Fattal HLDG, Sin EL Fil Jisr EL Wati Beirut, +Lithuania Elektronas AB, Kareiviu 6, LT 2600 Vilnius, 17 Luxembourg Sogel S. A., Rue de lindustrie 7, L-2543 Windhof, Malaysia Exact Quality Lot 24 Rawang Housing & Industrial Estate Mukim Rawang, PO No 210 48,000 Rawang Selangor Malaysia (603) Malta Kinds, 287, Republic Street, Valletta VLT04, 18 Morocco FMG depot TMTA Rue Chefchaouen, Oukacha Casablanca +69 Martinique Decius Absalon, 23 Rue du Vieux-Chemin, 97201 Fort-de-France, 15 Mauritius J. Kalachand & Co. Ltd., DBM Industrial Estate, Stage 11, P.O.B. 634 Plaine Lauzun, 84 10
Mexico Gillette Manufactura, S.A. de C.V./ Gillette Distribuidora, S.A. de C.V. Atomo No. 3 Parque Industrial Naucalpan Naucalpan de Juarez Estado de Mxico, C.P. 53370 01-800-508-58-00 Nederland Gillette Groep Nederland BV, Visseringlaan 22, 2288 ER Rijswijk, 0 800-88 Netherlands Antilles Rupchand Sons n.v. (rams), Front Street 67, P.O. Box 79 St. Maarten, Philipsburg 31 New Zealand Key Service Ltd., 69 Druces Road., Manakau City, 09 - 38 Nippon Gillette Japan Inc., Queens Tower, 13F 3-1, Minato Mirai 2-Chome Nishi-Ku, Yokohama 220-6013 Japan 045-00 Norge Gillette Group Norge AS, Nils Hansensvei 4, P.O. Box 79 Bryn, 0667 Oslo, 022-10 Oman ( Sultanate of ) Naranjee Hirjee & Co., 10 Ruwi High, P.O. Box 9, Muscat 113, Pakistan Gillette Pakistan Limited, Dr. Ziauddin Ahmend Road, Karachi 74200, Paraguay Paraguay Trading S. A., Avda. Artigas y Cacique Cara Cara, Asuncin, 21203350-48/46 Philippines Gillette Philippines Inc., Corporate Corner Commerce Avenues 20/F Tower 1, IL Corporate Centre 1770 Muntinlupa city 02-771071 02-06/-16
6022230_IRT_MN_S4-66 Seite 54 Mittwoch, 2. Februar 2005 12:37 12
Poland Gillette Group Poland Sp.Z o.o, ul. Domaniewska 41, 02-672 Warszawa, Portugal Grupo Gillette Portugal, Lda., Braun Service, Rua Toms da Fonseca, Torre G - 2 A, 1600-209 Lisboa, Runion Dindar Confort, Rte du Gymnase, Boite Postale 278, 97940 St. Clotilde, Romania Gillette Romania srl. Calea Floreasca nr. 133-137 et 1, sect 1, 714011 Bucuresti 01-2319656 Russia RTC Sovinservice, Rusakovskaya 7, 107140 Moscow, (095) 61 Saudi Arabia AL Naghi company AL Madinah road opposite to Fetihi center, Al Forsan P.O. Box: Jeddah 02 - Schweiz /Suisse/Svizzera Telion AG, Rtistrasse 26, 8952 Schlieren, 0844-10 Singapore (Republic of) Beste (S) Pte. Ltd., No. 6 Tagore Drive, # 03- 04 Tagore Industrial Building, Singapore 787623, (65) Slovakia Techno Servis Bratislava, Bajzova 11/A, 82108 Bratislava, (02) Slovenia Iskra Prins d.d. Rozna dolina c. IX/Ljubljana, 00
South Africa (Republic of) Fixnet After Sales Service, 159 Queen Street, Kensington South, P.O. Box 751770, Johannesburg 2094, Kensington South, 6765 St. Maarten Rupchand Sons n.v. (rams), P. O. Box 79, Philipsburg, Netherlands Antilles, 31 St. Thomas Boolchands Ltd., 31 Main Street, P. O. Box 5667, 00803 St. Thomas, US Virgin Islands, 0302 Suomi Gillette Group Finland Oy, P.O. Box 9, Niittykatu 8, PL 9, 02200 Espoo, 09-71 Sverige Gillette Group Sverige AB, Dept. Sweden, Stockholm Gillette Rsundavgen 12, Box 702, 16927 Solna, 020-21 Syria Ahmed Hadaya Company Hadaya building Ain Keresh Unisyria, P. O. Box 35002, Damascus, 963 011-231433 Taiwan Audio & Electr. Supplies Ltd., Brothers Bldg., 10th Floor, 85 Chung Shan N Rd., Sec. 1, Taipei (104), (886) 3283 Thailand Gillette Thailand Ltd., 175 South Sathorn Road, Tungmahamek, Sathorn, 11/1 Floor, Sathorn City Tower Bangkok 10520 (66) 2344 9191/ Exten. 9135 Tunesie Generale dEquipement Industr., (G.E.I.) 34 rue du Golfe Arabe, Tunis, 2000, 80
Turkey Gillette Sanayi ve Ticaret A.S., Polaris Is Merkezi, Ahi Evran Cad., No:1, 80870 Maslak, Istanbul, 0212-85 Ukraine Importbytservice-Ukraine Hlybotchytska str.53, Kyiv 380-44-417-24-15 United Arab Emirates The New Store LLC, Burjman shopping mall, Bur Dubai Dubai, +Uruguay Driva S. A., Marcelilno Sosa 2064, 11800 Montevideo, USA The Gillette Company Braun Consumer Service, 1, Gillette Park 4k-16, Boston, MA 02127-1096, 1- 800 -272 - 8611 Venezuela Gillette de Venezuela S.A., Av. Blandin, Centro San Ignacio Torre Coprnico, Piso 5 La Castellana, Caracas 0800-4455388 Yemen ( Republic of ) Saba Stores for Trading, 26th September Street, P. O. Box 5278, Taiz, 4-88 Yugoslavia BG Elektronik, Bulevar kralja Aleksandra 34, 11000 Beograd, 030
6022230_IRT_MN_S4-66 Seite 57 Mittwoch, 2. Februar 2005 12:37 12
Normal body temperature is a range. The following table shows that this normal range also varies by site. Therefore, readings from different sites should not be directly compared.
C 38.5 38
Normal ranges by site: Axillary1:
34.7 C 37.3 C 35.5 C 37.5 C 36.6 C 38.0 C 35.8 C 38.0 C
94.5 99.1 F 95.9 99.5 F 97.9 100.4 F 96.4 100.4 F
Oral1: Rectal1: ThermoScan2:
37 36.5 35.34.5
However, the range of normal also varies from person to person and fluctuates throughout the day. It is therefore important to determine your and your family members' normal temperature ranges. This is easily done using Braun ThermoScan. Practise taking temperatures on yourself and healthy family members to determine their normal temperature ranges.
Family Member Date 8:00 h 12:00 h 16:00 h
Note: When consulting your physician, communicate that the ThermoScan temperature is a temperature measured in the ear and if possible, note the individual's normal ThermoScan temperature range as an additional reference. In case of questions, call Braun Infoline.
1 Chamberlain, J. M. Terndrup, T. E., New Light on Thermometer Readings, Contemporary Pediatrics, March 1994. 2 Chamberlain, J. M., et al., Determination of Normal Ear Temperature with an Infrared Emission Detection Thermometer; Annals of Emergency Medicine, January 1995.
6022230_IRT_MN_S4-66 Seite 58 Mittwoch, 2. Februar 2005 12:37 12
How to use Braun ThermoScan
1. To achieve accurate readings, make sure a new, clean lens filter is in place before each measurement.
2. IRT 4020: Push the start button. IRT 4520: Push the I/O button. Then wait for the ready signal beep and the ready symbol in the display.
3. Fit the probe snuggly into the ear canal, then push the start button.
4. If the probe has been positioned securely into the ear canal during the complete measuring process, a long beep will signal the end of the measuring process. You can be assured to have taken an accurate temperature reading. The result is shown on the display.
If you take the temperature of another person, the ExacTemp light will be of help. It flashes during the measuring process as long as the probe is securely positioned, and lights up continuously when an accurate reading has been taken.
NN-E225mbbpq CDX-GT427UE DVP-K370 Altima-2005 RD 127 HL-730 Speed 7913 Edition 3 Kxtg6411RU Jetdirect 175X Versatis 155 Sony EVC3 SC101 Powermax 230 Array Classico WM5000 L1750SQ 6VIA3 62SX4R Pasadena CD71 Center WD-14126RD CDP-CX205 LSP-400R Keyboards KX-TS105 DS-114AT LQ-870 KDL-32S4000E WT-R851 ALL-IN-ONE TXL32D25E 66 SR-30 Officejet 7130 2929 FR Vivicam 3705 HT-SS1000 Server UB802 FM-8500 FW530C WIM 2160 Keyboard XM601 L14720 CQ-RD110 14 4 VCT-1500L TH-50PH9 APS57 FTB-200 MRW62E-s2 ZWI285 Servers Tamron 572D SRW2008MP 1 0 X7675 Pro 500 Motorola H620 Radeon X850 42PD6600 SGH-F200 Es500 Compact DZ-GX5040E DJ-X7E TX-P37x20E PV-GS35 7 7E OT-880A DD600 Tutorial Sansa E100 NN-A574sbbpq Server Zans710 BEQ700 MP4-player KAM200 WD8122CVD MAX-ZS990 B2430HD Speedtouch 510I EWS 1250 Bizhub 421 Fbii XL-2 26LC41 Desktop MX V-clic DVP3005 SPP-A2770 LT-42DG8BJ 900087 UX-H10 RB-1572 9 DCI GV-HIA6
manuel d'instructions, Guide de l'utilisateur | Manual de instrucciones, Instrucciones de uso | Bedienungsanleitung, Bedienungsanleitung | Manual de Instruções, guia do usuário | инструкция | návod na použitie, Užívateľská príručka, návod k použití | bruksanvisningen | instrukcja, podręcznik użytkownika | kullanım kılavuzu, Kullanım | kézikönyv, használati útmutató | manuale di istruzioni, istruzioni d'uso | handleiding, gebruikershandleiding
1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30 31 32 33 34 35 36 37 38 39 40 41 42 43 44 45 46 47 48 49 50 51 52 53 54 55 56 57 58 59 60 61 62 63 64 65 66 67 68 69 70 71 72 73 74 75 76 77 78 79 80 81 82 83 84 85 86 87 88 89 90 91 92 93 94 95 96 97 98 99 100 101