Hanna Instruments Phep
This accurate, inexpensive sensor is ideal for taking fast, simple measurements of solutions. The self-contained, miniaturized instrument has an operating temperature range of 0-50 degrees C, a large, easy-to-read liquid crystal display, and a manageable size and weight (85 g).
Part Number: HI98107
[ Report abuse or wrong photo | Share your Hanna Instruments Phep photo ]
Hanna Instruments Phep
User reviews and opinions
No opinions have been provided. Be the first and add a new opinion/review.
Dear Customer, Thank you for choosing a Hanna product. If you need additional technical information, do not hesitate to e-mail us at email@example.com. This instrument is in compliance with the directives.
HI 96108 Pocket-Sized pH Meter with ATC
Range 0.0 to 14.0 pH Resolution 0.1 pH Accuracy (@20C) 0.1 pH Calibration manual, 2 points Temperature compensation automatic, 0 to 50C (32 to 122F) Environment RH 95% max; 0 to 50C (32 to 122F) Batteries 3 x 1.5V alkaline Life approx. 150 hours of use Dimensions 150x30x21 mm (5.9x1.2x0.8") Weight 70 g (2.5 oz.)
HI HI HI HI HI 7004M 7007M 7010M 70300M 7061M pH 4.01 solution (230 ml) pH 7.01 solution (230 ml) pH 10.01 solution (230 ml) Storage solution (230 ml) Cleaning solution (230 ml)
This meter is guaranteed for six months against defects in workmanship and materials when used for its intended purpose and maintained according to instructions. This warranty is limited to repair or replacement free of charge. Damages due to accident, misuse, tampering or lack of prescribed maintenance are not covered. If service is required, contact the dealer from whom you purchased the instrument. If under warranty, report the model number, date of purchase, serial number and the nature of the failure. If the repair is not covered by the warranty, you will be notified of the charges incurred. If the instrument is to be returned to Hanna Instruments, first obtain a Returned Goods Authorization Number from the Customer Service department and then send it with shipment costs prepaid. When shipping any instrument, make sure it is properly packaged for complete protection. IST96108 12/04
Remove the protective cap. Do not be alarmed if white crystals appear around the cap. This is normal with pH electrodes and they dissolve when rinsed with water. Turn the meter on sliding the switch on the top. Immerse into solution up to the max immersion level. Stir gently and wait until the display stabilizes. After use, rinse the electrode with water to minimize contamination and store it with a few drops of storage (HI70300) or pH 7 (HI 7007) solution in the protective cap. DO NOT USE DISTILLED OR DEIONIZED WATER FOR STORAGE PURPOSES. Large fluctuations in readings could be due to lack of calibration, dry electrode or rundown batteries.
Recommendations for users Before using this product, make sure that it is entirely suitable for the environment in which it is used. Operation of this instrument in residential areas could cause interference to radio and TV equipment. The glass bulb at the end of the pH electrode is sensitive to electrostatic discharge. Avoid touching this glass bulb at all times. Any variation introduced by the user to the supplied equipment may degrade the instrument's EMC performance. To avoid electrical shock, do not use this instrument when voltages at the measurement surface exceed 24 Vac or 60 Vdc. To avoid damage or burns, do not perform any measurement in microwave ovens.
Immerse the tester up to the maximum immersion level in pH 7 buffer (HI 7007) Allow the reading to stabilize; using a small screwdriver adjust the pH7 calibration trimmer to read 7.0. Rinse the electrode with water and immerse up to the maximum level in pH 4 (HI 7004) or pH 10 (HI 7010). Allow the reading to stabilize and adjust the pH4/10 calibration trimmer to read 4.0 (or 10.0). Calibration is now complete.
ALWAYS USE FRESH BUFFERS FOR CALIBRATION.
When the meter cannot be switched on or the display fades, pull out the battery compartment and replace all three 1.5V batteries, while paying attention to their polarity. Batteries should only be replaced in a safe area using the battery type specified in this instruction manual.
pHep is a registered Trademark of "Hanna Instruments"
R6008HS LV2393 LE-40R81B Ix IC-737A 32PW9767 MS-74ML Griso 8V Silverado 2005 DP889 B215A SD200 DVD-P370-EUR Discovery MYH10 10826 800-2 RA Mini DV RX505 KX-TCA94EX DTB-9401V RM-EZ2T HDR-UX19E F60860 Xdrive 35I Urc-4220 DP9210 TP329 SPF-85V Ellipse AV-pack 207 Emtec C250 Specs 10505 Korg I2 AVR-487 DPL2000 HT-SB300 QW-1044 DV-420V-S CP-29Q54 DG60-112 7720Z MD-105TX Limited LTD1 NN-SD277S GV-A500 RR-QR240 Seiko 7T62 FAX-L200 PSR-3000 Smoochie PUP IC-271H AX-630 CDM7075 Cleaner VP-D305I R-210A 32PF9541 Street XV-5080 PS42B430p2W Radiance XS DPF-D1020 Multiquick Casio 4334 TDM-IP20 TK-3302 Impressa X9 42PMA400E Protocol LQ-1170 Wrtp54G-ER SA1115 Gunship Cl2939 Essential SE SR-S20FTC Exai5180 Review EDE419M LW053 3260C GT-S7220 KB-155 Moto Z8 Flymo L300 MHC-BX5 SWM-745 D-540 Zoom Canon I950 Onetouch HP 48GX EMP-TW10H DM-80 C61R2AEE Software CTK-481 Beosound 9000 VOD1021 CDA-9827
manuel d'instructions, Guide de l'utilisateur | Manual de instrucciones, Instrucciones de uso | Bedienungsanleitung, Bedienungsanleitung | Manual de Instruções, guia do usuário | инструкция | návod na použitie, Užívateľská príručka, návod k použití | bruksanvisningen | instrukcja, podręcznik użytkownika | kullanım kılavuzu, Kullanım | kézikönyv, használati útmutató | manuale di istruzioni, istruzioni d'uso | handleiding, gebruikershandleiding
1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30 31 32 33 34 35 36 37 38 39 40 41 42 43 44 45 46 47 48 49 50 51 52 53 54 55 56 57 58 59 60 61 62 63 64 65 66 67 68 69 70 71 72 73 74 75 76 77 78 79 80 81 82 83 84 85 86 87 88 89 90 91 92 93 94 95 96 97 98 99 100 101