Here you can find all about Pentax R-326EX like manual and other informations. For example: review.
Pentax R-326EX manual (user guide) is ready to download for free.
On the bottom of page users can write a review. If you own a Pentax R-326EX please write about it to help other people. [ Report abuse or wrong photo | Share your Pentax R-326EX photo ]
Pentax R-326EX Digital Camera, size: 2.9 MB
Pentax R-326EX Ptl Software
Pentax R-326EX Brochure
Pentax R-326EX Quick Reference Guide
Pentax R-326EX Quick Reference Guide Powertopolite
User reviews and opinions
|suebaby41||6:50pm on Friday, October 15th, 2010|
|I picked up my K100D a few months back after ... This has been the best camera I have purchased in my life. I was looking at other comparable models.|
|Johnlee||5:55pm on Monday, September 6th, 2010|
|Here today, gone tomorrow This camera has an awesome image stabilization feature. Great Camera Well, I have had this camera for more than 2 years now and I can say it is GREAT.|
|katopen||10:42am on Sunday, August 15th, 2010|
|I think Pentax K100D Super ia a Popular-Class Lens-Interchangeable Digital SLR Camera Featuring Innovative Shake Reduction System and a Host of Advanc...|
|Randy Knutson||4:29pm on Tuesday, August 3rd, 2010|
|The camera was great up until it stopped working, about 5 months after purchase. I have been quite pleased with my purchase thus far. This comment is for beginners to DSLR photography.|
|dreamdragon||7:29pm on Thursday, April 15th, 2010|
|Compared to other similar cameras what I like most in Pentax - Natural & reach colors - much better than Canon produce. Build quality is very good .|
Comments posted on www.ps2netdrivers.net are solely the views and opinions of the people posting them and do not necessarily reflect the views or opinions of us.
Total Surveying Solutions
Electronic Quick Reference Guide Total Station
R-322NX R-323NX R-325NX R-335NX R-315NX
R-322EX R-323EX R-325EX R-335EX R-315EX R-326EX
The reflectorless function is applied to R-322NX, R-323NX, R-325NX, R-326EX, R-335NX, R-315NX
Basic Procedures for R-300X Series
PENTAX Industrial Instruments Co., Ltd.
2-5-2 Higashi-Oizumi / Nerima-ku, Tokyo 178-0063, Japan Tel. +/ Fax +E-mail: email@example.com Website: www.pentax.co.jp/piic/survey www.pentaxR300X.com
Electronic Total Station
Quick Reference Guide Basic Procedures for R-300X series
R-322NX, R-323NX, R-325NX, R-335NX, R-315NX, R-322EX, R-323EX, R-325EX, R-335EX, R-315EX, R-326EX
The description concerning the reflectorless function in this guide is applied to R-322NX, R-323NX, R-325NX, R-326EX, R-335NX, R-315NX.
General Instruction Manuals Precautions regarding safety Warning Usage Precautions 1 Basic Operation 1.1 Removing the Battery 1.2 Attaching the Battery 1.3 Turning the Power On and Off 1.4 Centering and Leveling 1.5 Target Sighting (Focusing) 1.5.1 Auto-Focus 1.5.2 Power Focus 1.5.3 Manual Focus 1.5.4 Continuous AF mode 1.6 Angle Measurement 1.6.1 Horizontal Angle 1.6.2 Vertical Angle 1.7. Distance Measurement 1.7.1 Select your target 1.7.2 Distance Measurement 1.7.3 Changing Target constants 1.7.4 Input Temperature and Atmospheric pressure 1.7.5 Laser pointer 1.7.6Adjusting Laser Plummet brightness 1.7.7 Adjusting LCD contrast 1.7.8 Adjusting Illumination brightness 2 Changing Instrument Settings 3 Basic Field Checking Procedures 3.1 Electronic Vial 3.2 Laser Plummet 3.3 Laser Pointer 3.4 Warning and Error Messages 3.5 Atmospheric Correction 4 Specifications Notice to the user of this product
Instruction Manuals Quick Reference Guide is intended to provide a quick reference in the field. For ease of use in the field, the following Quick Reference Guide booklets are provided in the carrying case. 1. Basic procedure 2. PowerTopoLite for R-300X series, Operating procedure The complete instruction manuals are contained on the CD that is attached to each R-300X. This guide uses the symbol xN as an expression of repeating times of key operation. For example. ESC x2 means that [ESC] key is pressed two times. The symbol +expresses that multiple keys are pressed simultaneously. PRECAUTIONS REGARDING SAFETY Before using this product, be sure that you have thoroughly read and understood the instruction manual that is included in the attached CD-ROM to ensure proper operation.
Solar Observation Never view the sun directly using the telescope as this may result in loss of sight. Laser Safety R-300NX is a class-IIIa (3R) Laser product. Avoid direct eye exposure. R-300X without NX is a class-II (2) Laser product. Do not stare into laser beam. Electro-Magnetic Compatibility (EMC) This instrument complies with the protection requirement for residential and commercial areas. If this instrument is used close to industrial areas or transmitters, the equipment can be influenced by electromagnetic fields.
Risk of Explosion Do not use this product in a location where there is coal dust, or near flammable material as there is a risk of explosion.
Target Constant Confirm the Target Constant of the instrument before measurement. Reflectorless and Reflecting sheet The reflectorless measurement range may vary depending on the target and surrounding brightness. In case the reflectorless measurement results in low accuracy, perform the distance measurement by Reflector sheet or Prism. (R-322NX, R-323NX, R-325NX, R-335NX, R-315NX,) Battery & Charger Use the battery charger that is suitable to the battery you are using. If water should happen to splash on the instrument or the battery, wipe it off immediately and allow it to dry in a dry location. Auto focus The Auto focus may not function under every condition depending on brightness,contrast,and the shape and size of the target.In such a case, use the Power Focus buttons or the manual focus ring. Atmospheric correction During surveys for which the survey precision or atmospheric measurement method is defined, measure the atmospheric temperature and pressure separately and enter those values rather than using the Automatic Atmospheric Correction function.
1. Basic Operation
1.1 Removing the Battery (1) Rotate the knob above the battery pack counter-clockwise. (2) Lift up the battery pack and remove it from the instrument. 1.2 Attaching the Battery (1) Place the channel on the bottom of the battery pack, onto the protrusion of the instrument and push the battery pack down into place. (2) Turn the knob clockwise. 1.3 Turning the Power On and Off To set power on : To shut down : Electronic vial screen
NOTE: The power is automatically turned off after 10 minutes of inactivity (Factory default setting). 1.4 Centering and Leveling (Laser plummet and Electronic vial) To activate electronic vial : To activate laser plummet :
(for the first time after power on)
or: To deactivate :
NOTE: The laser plummet and the electronic vial are always activated at the same time.When the power is switched on, however, only electronic vial is activated (Factory default setting). To change the power-on activation of laser plummet, refer to 2.3 Instrument setting itemsF.
1.5 Target Sighting (Focusing) 1.5.1 Auto-Focus Aim using the telescope collimator, and press NOTE: Place the target near the reticle center.The AF sensor is located around the horizontal line of the reticle. 1.5.2 Power focus If the AF failed or if you need to adjust the focus, use the Power focus knob. To focus on a closer object : Rotate Power Focus knob clockwise To focus on a farther object : Rotate Power Focus knob counterclockwise NOTE: Focus speed can be controlled by the rotation angle of the knob: Low speed: when rotated by approx. 5 degrees Middle speed: when rotated fully by approx. 10 degrees High speed: after one second of time has passed in the middle speed position 1.5.3 Manual focus Rotate the focus ring manually: 1.5.4 Continuous AF mode To activate the continuous AF: for 2 seconds
NOTE: Track the target by keeping the target close to the center of the reticle as much as possible. To quit: or rotate the Power focus knob :
NOTE: In case the focus does not change for one minute, the continuous AF mode will also terminate automatically.
1.6 Angle Measurement 1.6.1 Horizontal Angle Set the screen MODE A :
0 SET F3 HOLD F3
MODE A screen
Control keys for measuring horizontal angle: To set the angle to 0 To hold the angle To release HOLD To input an angle : : : :
0 SET F3 MODE F5 HOLD F3 MODE F5 ANG SET F2 F4 SELECT F5 ENT
input value by using: To read clockwise angle : 1.6.2 Vertical angle To display vertical angle :
F1 MODE F5
F2 ANG SET F2
To change permanently the combination of values displayed in MODE A: ESC x2 SELECT x3 SELECT +
F2 F5 F4 F5
ENT F4 MODE F5
To read the slope %
ANG SET F2
1.7 Distance Measurement Set the screen MODE A:
1.7.1 Select your target Select target type (measurement mode): (Reflecting sheet)
NOTE: The selected target is maintained until next time you change. 1.7.2 Distance measurement For a single shot measurement:
When Long Range Mode is displayed
NOTE: Long Range Modeis displayed only when measuring with higher laser power (in reflectorless measurement with the instrument setting REF.LESS RANGEset as LONGand Warning Messageset as ON).
For tracking measurement
NOTE: The number of shots can be defined.The default is one time The measuring modes activated by the above operations. can be also changed.
NOTE: Quick mode is the function to shorten the initial measurement time using prism and reflector sheet.To set the mode, use direct command number .When the meas key is pressed during quick mode, <-- --> is displayed on the screen. 1.7.3 Changing Target constants The default constants are: Reflecting sheet : 0mm Prism : -30mm Reflector-less : always 0mm Before changing the constants, set Target Constant in the Initial Setting to INPUT mode: ESC SELECT x2 x2 ENT x3 +
F4 F5 F4
To change Prism constant: input value by using : ,
SELECT F5 ENT
CORR F4 F4
To change Sheet constant: input value by using:
1.7.4 Input Temperature and Atmospheric pressure The default atmospheric correction mode is Automatic. Before manual input, change the default mode to ATM INPUT : +
SELECT F5 CORR F4 F4 F4
To Input temperature
x2 , 0
input value by using:
To Input atmospheric pressure:
MODE F5 CORR F4 F4
x3 , 0
input value by using: 1.7.5 Laser pointer To activate Laser pointer : To quit Laser pointer :
LD POINT F2 LD POINT F2
NOTE: The laser pointer is kept activated until it is deactivated by the above procedure. 1.7.6 Adjusting Laser Plummet brightness
1.7.7 Adjusting LCD contrast +
LCD F4 F1
1.7.8 Adjusting Illumination brightness +
2. Changing Instrument settings
You can change the instrument settings by HELP menu or by inputting 007 code. 2.1 Help menu While the screen is in MODE A or MODE B, +
Select a desired item by
Change the setting
NOTE: Some items have sub-menus where the selecting procedure by using F1 - F4 is again repeated. 2.code While the screen is in MODE A or MODE B,
Input 007 code by using 0
NOTE: Some items have sub-menus where the selecting procedure by using F1 - F4 is again repeated. 2.3 Instrument setting items See chart on page 15
007 code 401 PRISM CONST SHEET CONST SHOT CONT SHOT INPUT
HELP menu list TARGET CONST
ATM CORR MEAS. MIN DISP SHOT COUNT
PRIM.MEAS KEY SEC.MEAS KEY AUTO MEAS. PRIORITY DISP
CRV/REF CORR MIN UNIT ANG. V. ANG. STYLE DIST.BUZ QUAD BUZ AUTO OFF EDM OFF ILLU.OFF MEAS.SIGNAL PRIORITY SELECT
LD PLUM. TILT DISP. TILT DISP.UNIT RANGE MESSAGE SETUP TEMP.UNIT PRESS UNIT BAUD RATE DATA LENGTH PARITY BITS STOP BITS SIGNAL CONTROL XON/XOFF THROUGH COMMAND
COMP AXIS LD PLUM.&E.VIAL
QUICK MEAS ATM UNIT
DIST.UNIT ANG.UNIT SET UP COM.
Default -30mm 0mm AUTO COARSE 1 time 01 times 0.14 FINE Z.0 ON OFF 10 MIN 3 MIN 3 MIN MARK MEAS SHOT TRACK CONT NIL HA/HD/VD 3 Axis (2/3model) or 2 Axis OFF (when Power On) OFF COARSE NORMAL ON (when Range is LONG) EACH TIME (w/ Power ON) OFF Centigrade hPa m DEG NIL 1 ON ON NIL
Other options 0mm, INPUT INPUT ATM INPUT, ppm INPUT, NIL FINE 3 times, 5 times, INPUT (input) 0.2, NIL COARSE H.0, COMPASS OFF ON 20 MIN, 30 MIN, NIL 5 MIN, 10 MIN, NIL 5 MIN, 10 MIN, NIL VALUE MEAS CONT, TRACK SHOT, TRACK CONT TRACK SHOT, MEAS CONT, MEAS SHOT MEAS., TRACK HA/VA/SD, HA/VA/HD/SD/VD 2 Axis (2/3 model), 1Axis, , NIL ON (automatically ON) ON FINE LONG OFF PERMANENT ON Fahrenheit mmHg, inchHg ft, ft+inch DEC, GRD, MIL 2400, 4800, EVEN, ODD 2 OFF OFF a, b, c, d, e,f
3. Basic Field Checking Procedures
Checks and Adjustments should be performed before and during measurement. 3.1 Electronic Vial To display vials:
(from measurement screen)
Check if the bubble stays at the center of each vial, when rotating the instrument by 180. To adjust the vials: + instruction on the screen operate according to the
NOTE: Press these two keys for 1 second longer, then release key first.
3.2 Laser Plummet To activate the laser plummet: (for the first time after Power on) or
Check if the laser spot on the ground stays at the same position when rotating the instrument around the vertical axis. To adjust the laser plummet: Contact your local dealer 3.3 Laser Pointer To activate the laser pointer: or (from measurement screen)
Check if the projected laser spot points at the same position that is aimed by the center of the cross-hair line of the telescope. To adjust the laser pointer: Consult your local dealer.
3.4 Warning and Error Messages Warning Message
Out of tilt range
Displayed when the instrument is tilted beyond the vertical compensation range (3) in case 1 axis or 2 axis automatic compensation is selected. This message may be temporarily displayed if the instrument is turned too fast.
What to do
Re-level the instrument. Repair is needed if the message is displayed when it is properly leveled.
The input data exceeds the allowable Press the [ESC] key and enter the correct data. range. This message is displayed if a long Select the correct target mode. distance which is a far beyond measurable distance of R-300X series is measured with a wrong target mode. Please select a correct target then measure. If a wrong target is selected, a correct distance cannot be measured. Select a longer point, The measurement distance is less than 1.5m in Reflector sheet mode. or use a tape measure. The measurement distance is less than 1.5m in Prism mode. Under too strong sun light. Unstable light value owing to shimmer or obstacles. Reflector sheet, Target and Prism do not face the instrument. Reflector sheet,Target and Prism are not correctly sighted. Measurement range is over in Reflectorless mode. Sufficient signal does not return by sighting sharp edge etc. at Reflectorless mode. The Date Clock is powered by the built-in lithium battery. The lithium battery needs to be replaced in five years. Change the object that has much better reflectivity, or use a reflecting sheet, or wait until the sun activity has weakened.
Target is too close.
Li-batt.voltage is low.
Have the lithium battery replaced by the dealer from whom the instrument was purchased.
Error Message Meaning
ERROR!! EDM ERROR Distance measurement system problem 04 -05, 34-39, 50-53 ERROR!! ETH ERROR 70-76 Angle measurement system problem
Turn the power off, and then turn on again. Repair is needed when the message appears consistently.
Memory problem ERROR!! MEMORY ERROR 19
ERROR PS DATA of EDM Problem of the internal EDM parameters ERROR P DATA of EDM
ERROR ETH DATA
Problem of the internal ETH parameters
3.5 Atmospheric Correction The speed at which light travels through the air varies depending on the temperature and atmospheric pressure. The R-300X series is designed to measure distances at the speed of light in order to measure accurately, atmospheric correction needs to be used. The instrument is designed to correct for weather conditions automatically if the temperature and pressure are input. Correction is then carried out based on the following formula. Calculation formula: K= (276.26713 - 78.565271 P ) x10-6 273.14941 + t K: Atmospheric Correction Constant P: Atmospheric pressure (hPa) t: Temperature(C) Distance after Atmospheric Correction D = Ds (1+K) Ds: Measured distance when no Atmospheric Correction is used.
Reflectorless items are applicable to NX models only.
R-322NX R-323NX R-325NX R-335NX R-315NX R-322EX R-323EX R-325EX R-335EX R-315EX R-326EX Telescope Magnification 30 x Resolving power 3 Field of view 2.6% (1 30) Minimum focus 1.0m Focus Auto-focus / Power Focus / Manual Manual Distance measurement Visible Laser: Class II (2) Laser Class Visible Laser: Class II (2) / Class IIIa (3R) - Long range mode in Reflectorless (NX models only) Measurement range (Good condition) Reflectorless 1.5m - Normal range mode: 90m / Long range mode 200m (NXmodels only) Reflector sheet 1.5m-600m (800m) Mini prism 1.5m-1100m (1600m) 1P 1.5m-3400m (4500m) 1.5m-3000m (4000m) 1.5m-2000m (2800m) 3P 200m-4500m (5600m) 200m-4000m (5000m) 200m-2800m (3500m) Accuracy Prism (2+2ppm x D)mm (3+2ppm x D)mm Reflector Sheet Reflectorless (5+2ppm x D)mm (5+3ppm x D)mm At Auto-Atm.Correction Prism, Ref.sheet: (2 +10ppm x D)mm / (3+10ppm x D)mm Reflectorless: (5 + 10ppm x D)mm (NXmodels only) Minimum count Fine mode (0.1mm) Normal mode (1mm) Track mode (1cm) Measuring time Repeat meas. Normal: Prism, Ref.sheet 2.0sec.(1mm) Normal: Prism, Ref.sheet 1.2sec. (1mm) Track: Prism, Ref.sheet 0.4sec.(1cm) *Quick mode Initial meas. Normal: Reflectorless 2.3sec.(1mm) Normal: Prism, Ref.sheet 2.5sec. (1mm) (NXmodels only) *Quick mode Angle measurement Measuring method Absolute rotary encoder Detection 2 sides Minimum count 1 (2cc) / 5 (10cc) selectable Accuracy(ISO 17123-3) Compensator Triple axis Dual axis Tangent screw 2 speed 1 speed Sensitivity of vials Plate level (electronic) 30/1div. Circular level 8/2mm Plummet Visible Laser, 0.8mm (instrument height 1.5m) Base Detachable Shifting Fixed Detachable Water resistant IP56 Ambient temperature -20C ~ +50C / -4F ~122F (Working range) Tripod thread 5/8 x 11 35mm x 2 5/8 x 11 Dimensions/Weight Instrument 177(W) x 343(H) x 177(L)mm Weight (incl. Battery) 5.7kg 5.5kg 5.7kg Carrying case 268(W) x 442(H) x 465(L)mm/3.9kg Battery pack BP02 Power source Ni-MH (Rechargeable)(4300mAh) , DC6V Operation time Continuous Approx. 5 hrs (ETH+EDM),12 hrs (ETH) with Approx. 2.2 hrs of charging time Weight Approx. 380g
R-322NX R-323NX R-322EX R-323EX Charger BC03 and AC Adapter AC01 Input voltage (AC01) AC 100~240V (AC01) Output voltage (BC03) DC7.5V (BC03) Weight 280g Internal Memory Coordinates data 20000 points Date Clock NX MODEL Yes EX MODEL Yes
16000 points Yes No
12000 points No
NOTE: * The measurement range may vary by measurement conditions. * Normal conditions: 20km visibility with slight shimmer. * Good conditions: 40km visibility, overcast, no heat, no shimmer and moderate wind. * Reflector sheet: PENTAX genuine Reflector sheet (5cm x 5cm). * Quick mode, which is effective only under Normal mode (1mm) setting, functions with Prism and Reflector Sheet. It is incorporated in all X series models and effective up to 500m. * When Quick Mode is on, the EDM accuracy using Prism and Reflector Sheet is (3+2ppm x D)mm. When Automatic Correction is activated in Quick Mode, the EDM accuracy is (3+10ppm x D)mm * EDM Measuring time varies according to distance to be measured and conditions of the environment. * Reflectorless: - The measurement range and accuracy of Reflectorless are based on the condition that laser beam is emitted perpendicular to the white side of the Kodak Gray Card. There is a possibility that the range may vary when the target does not satisfy the conditions above at survey work. - The measurement range may be influenced by the shape of the target and its environment. - The measurement range at TRACK mode is over 5m. * The operating time becomes shorter under the low temperature, due to the temperature dependence of the battery performance. * The points that can be recorded in one job file are maximum 2,000 measuring points (XYZ). The user can make up to 50 job files. Job history can be managed thanks to date clock recorded in each job file.
NOTICE TO THE USER OF THIS PRODUCT
To assure compliance with the Safety standard 21 CFR, Chapter 1. Subchapter J. The U.S. bureau of Radiological Health requires the following information to be provided to user.: It can be dangerous to look into the beam with optical equipment such as binoculars and telescopes.
1. Specifications of Laser Radiation A) The EDM module of the R-300X produces a visible light beam, which is emitted from the telescope objective lens and the center hole of the instrument base plate.The R-300X is designed and built to have a laser diode radiating at 620-690nm. B) Radiant power The R-300X is designed and built to radiate a maximum average radiant power of 4.75mW (0.95mW for the model without NX) from the telescope,and 0.95mW from the center hole of the base plate.The user may be subject to this radiation as a beam while operation until such time that the instrument is turned off.
2. The following labels are affixed to and must remain attached to this laser product. A) The following Certification label is located near the Plate level: This laser product is complied with the provisions of 21 CFR 1040. 10 and 1040.11. For a Class II laser product. Or for R-300 NX models: This laser product is complied with the provisions of 21 CFR 1040. 10 and 1040.11. For a Class IIIa laser product. B) Caution label is located near the exit aperture: AVOID EXPOSURE. Laser radiation is emitted from this aperture.
C) Warning logotype is located on the surface of the telescope: CAUTION LASER RADIATION DO NOT STARE INTO BEAM Or for R-300 NX models: DANGER LASER RADIATION AVOID DIRECT EYE EXPOSURE D) Warning label is Located near the exit aperture.
3. Caution to maintain the safety in compliance with the standard A) To maintain the safety standard, refrain from any operation, maintenance, or adjustment other than described in this instruction manual. B) Operation, maintenance or adjustment other than those specified in this instruction manual may result in hazardous radiation exposure. C) Maintenance and repair not covered in this manual must be done by an authorized Pentax dealer. D) The Laser beam emission by the Distance measurement can be MEAS terminated by pressing F1 key. E) Pressing keys can terminate the laser beam emission F2 by the laser pointer. F) The Laser beam emission by the Laser plummet can be terminated by pressing key.
AVOID EXPOSURE Laser radiation is emitted from this aperture
Warning label Laser beam is transmitted from this aperture
R325EX R325NX R335EX R335NX R315EX R315NX R322EX R322NX R323EX R323NX R326EX
Laser radiation is emitted from this aperture
Laser beam is transmitted from this aperture
For R-300NX model
AVOID DIRECT EYE EXPOSURE
CLASS 3R LASER PRODUCT
For R-300EX model
R GE DAN
DO NOT STARE INTO BEAM CLASS 2 LASER PRODUCT
SE OIDEXPO ER LAS
ION AT E DIECT EY DUCT DIR R RA SUREPRO
For North America Certification
2-5-2 Higashi Ozumi Nerima-ku. Tokyo 178-8555 Japan This Laser product is complied with the provisions of 21 CFR 1040-10 and 1040-11 For a class IIIa product ASSEMBLED BY PENTAX VN Co. LTD Sai Dong B Industrial zone. Gia Lam. Hanoi Vietnam
2-5-2 Higashi Ozumi Nerima-ku. Tokyo 178-8555 Japan This Laser product is complied with the provisions of 21 CFR 1040-10 and 1040-11 For a class II product ASSEMBLED BY PENTAX VN Co. LTD Sai Dong B Industrial zone. Gia Lam. Hanoi Vietnam
LED is turned on at the time of emission
LASER RADIATION - DO NOT STARE INTO BEAM
CLASS II LASER PRODUCT
Laserclass II, conform FDA 21 CFR Ch. 1 1040
LASER RADIATION - DO NOT STARE INTO BEAM OR VIEW DIRECTLY WITH OPTICAL INSTRUMENT INTO SUNLIGHT
CLASS IIIa LASER PRODUCT
Laserclass IIIa, conform FDA 21 CFR Ch. 1 1040
620-690 nm/4.95mW max.
PENTAX Industrial Instruments Co., Ltd. 2-5-2 Higashi-Oizumi Nerima-ku, Tokyo 178-0063, Japan Tel. +Fax +E-mail: firstname.lastname@example.org Website: www.pentax.co.jp/piic/survey
Japan Surveying Instruments Manufacturers Association
Member symbol of the Japan Surveying Instruments Manufacturers Association representing the high quality surveying products.
022005 First edition TS01401
Printed in Belgium
21FU1RK ZWC85050W DSP 244 N6200 IC-2GXA 7010IX ST-SE500 Srteg10 Xv-n650 P-140-P-140S MS-1924W LDS4821ST LA1000T AV220G NW-S705F CD-SW200 SE7451B-37 TS-H552D XL550U AR-90 Temporis MAX V1000 GO 6430 DC-228 Motorazr V3R Neonode N1 Mini 3 RM313P Pctv 400I DMC-LC1 A-3300 A096UQ Up3 Command Broshure 8 0 6241-W CDE-9842 VGN-Z11xn B Kh 1166 1222FX CM-300 AT-897 TX-SA806X HL-5240L TH-37PW5 Review ICF-M55 P4B533-M E4300 SU9380 42PD5200 Keyboard Optio P80 Sparc 16V Spellmate 700 CJ-KS50 JT-P770 Z5500D Vivicam 3330 C2118 FHB500 AEG-electrolux 60 DES-1004 4000N Revolution HTS3366 98 DCR-PC330 Microtower PC DSC-T90 T Pv-dv102 Cayenne PX710W EH-TW4400 F1225 5604RK 15LW1R AFM35 DT50 R KDC-PSW9524 Rhapsody 6 PRO N4L-vm DH L204WT-BF B3312 P1120 4X4-2007 PMD-DOK2 DFW50 Dmcfz28 SK-T2000 Photosmart 1000 I-45CV HD DVR KR-350 63 G SNB6500 00 Fdrv-90E 600SC RSP-966 Motolivn D711 LCG3000
manuel d'instructions, Guide de l'utilisateur | Manual de instrucciones, Instrucciones de uso | Bedienungsanleitung, Bedienungsanleitung | Manual de Instruções, guia do usuário | инструкция | návod na použitie, Užívateľská príručka, návod k použití | bruksanvisningen | instrukcja, podręcznik użytkownika | kullanım kılavuzu, Kullanım | kézikönyv, használati útmutató | manuale di istruzioni, istruzioni d'uso | handleiding, gebruikershandleiding
1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30 31 32 33 34 35 36 37 38 39 40 41 42 43 44 45 46 47 48 49 50 51 52 53 54 55 56 57 58 59 60 61 62 63 64 65 66 67 68 69 70 71 72 73 74 75 76 77 78 79 80 81 82 83 84 85 86 87 88 89 90 91 92 93 94 95 96 97 98 99 100 101