Teco CA 240
Here you can find all about Teco CA 240 like manual and other informations. For example: review.
Teco CA 240 manual (user guide) is ready to download for free.
On the bottom of page users can write a review. If you own a Teco CA 240 please write about it to help other people. [ Report abuse or wrong photo | Share your Teco CA 240 photo ]
Teco CA 240, size: 75 KB
Teco CA 240
User reviews and opinions
|nicolasecarnot||3:04pm on Wednesday, September 8th, 2010|
|Speed If you want to add speed to your PC then this is the thing to buy. Have your OS on this and put all programmes on a separate 1TB drive. Awesome performance I got one of these (OCZ Vertex 2 120GB) for my 15" Macbook Pro, to replace a 7200 RPM Seagate drive. OZC SSD Vertex Series This is by far the best upgrade i have made, it is not my first OZC product and it will not be the last. Super fast Boot time.|
|jeuf||7:54am on Tuesday, August 24th, 2010|
|none... beside the cons... other then that.. one thing "just buy it" :P Fast, HDtach run as window xp on window 7 ult saids its 240 brust read speed,... Not a bad PSU by any stretch, but definitely not compatible with my build. Summarizing the storagereview article: Synthetic benchmarks show a large speed difference between the original vertex 2 (32nm) and the newer vertex 2 ...|
|Qalibur||7:02am on Sunday, May 9th, 2010|
|Quick and Fast but way too small Purchase this SSD to use in my new computer build. Works great and setting it up was really easy. UNRELIABLE Check the ocz ssd forum to see the problems associated with this drive and other products from this manufacturer. Great improvement over HDD Compared to the prev. 120GB hard drive in my laptop, this drive is a big improvement in booting.|
|colinmain||11:10am on Saturday, March 27th, 2010|
|Amazing Product This is my first SSD and I am very impressed. I intentionally paid a little more for one that had a higher read/write rate.|
|startling||1:36pm on Friday, March 26th, 2010|
|I came in here while doing a RMA on this drive and saw all these reviews of it dying within the first year.|
Comments posted on www.ps2netdrivers.net are solely the views and opinions of the people posting them and do not necessarily reflect the views or opinions of us.
MAR 2 7200b
1268 N. Lakeview Ave., Anaheim, CA 92807, U.S.A. Phone: (714) 693-7788 Toll Free: 1-800-222-9880 ~~~~~Fax: (714) 693-3838 www.tecodia2.com Email:tecodiag~tecodiag.com
Updated 510OK summary Teco CX3 Reagent Set for SYNCHRON CX Contact Name: Jian Vaeches Phone Number: 714-693-7788 ext. 131 Prepared Date: 03/20/2006
1268 N. Lakeview Ave., Anaheim, CA 92807, U.S.A. Phone: (714) 693-7788 Toll Free: 1-800-222-9880
Fax: (714) 693-3838
Device Name The device trade names and common/classifications name are:
Device Trade Name Teco CX3 Reagent Set for SYNCHRON CX System Common/Classification Name BUN, CALCIIM, REATININEGLUCOSE REGENTS
Address and Registration
The address and registration number of the manufacturer site for Teco CX3 Reagent Set for SYNCHRON CX System: TECO Diagnostics 1268 N. Lakeview Ave. Anaheim, CA 92807, U.S.A. FDA Registration # 1832216
Teco CX3 Reagent Set has been classified as Class II with Product Code ~lo, JFt CGX, CGA "in vitro" diagnostics reagent set having the classification number: 21 CFR. 862.1770, 21 CFR 862.1145, 21 CFR 862.1225, 21 CFR 862.1345. This is the description available from the classification names listed in the "CDRH Home Page- Listing Database."
Predicate Device Information
The predicate device is Teco BUN Liquid, Teco Calcium Color, Teco Creatinine and Teco Glucose Liquid Reagent. The 510 (K) approval letter is provided in Appendix I. 510(K) #: K981106, K864741, K880629, K863926. Approval Date: 04-16-1998, 01-09-1987, 03-23-1988, 11-13-1986
Labeling and Intended Use
Draft labels and Instructions for use can be found in appendix A.
Teco CX3 Reagent Set for SYNCHRON CX is intended for the quantitative determination of BUN, Calcium, Creatinine and Glucose in serum on Beckman CX3 System.
Device Description and Comparison
The Teco CX3 Reagent Set for SYSNCHRON CX designed for use on the Beckman CX System. The Reagent Set includes BUN, Calcium, Creatinine and Glucose. Method Comparison of Teco CX3 Reagent Set for SYNCHRON CX to Teco BUN Liquid, Teco Calcium Color, Teco Creatinine and Teco Glucose Liquid following the guidelines of NCCLS Guideline EP9-A2 was conducted. Comparison with predicate: Similarities Intended use. Sample Handling Subsystem. Operational Environment Performance.
Differences Predicate Device Packaging Size BUN
Calicium Creatinine Glucose Analyzer Components BUN Tris Buffer 100 mmol 2-Oxoglutarate 5 mmol Urease >20,000 U/L GLDH > 1500 U/L NADH 0.25 mmol/L Predicate Device Candidate Device Urease >592 U/mi
5 x 25 ml & 5 x 5 ml
4 x 120 ml 4 x 120 ml 4 x 120 ml General Chemistry Analyzers
500 ml 2000 ml 500 ml SYSCHRON CX3
O-Cresolphthalein Complexone 0.14 mM 8-Hydroxyquinoline 13 mM Diethylamide 363 mM
Arsenazo III 0.15 mml/L
Picric Acid 10mM Sodium Hydroxide 240mM
Picric Acid 50mM Sodium Hydroxide 0.188 mM Glucose Oxidase 590 u/mI Ethanol 10% Potassium Iodide 0.04 mM
Glucose Oxidase 15 u/ml Peroxidase 1.2 u/ml 4-Aminoantipyrine 0.38 mM p-Hydroxybenzene Sulfonate 10 mM
1268 N. Lakeview Ave., Anaheim, CA 92807, U.S.A. Phone: (714) 693-7788 Toll Free: 1-800-222-9880 Fax: (714) 693-3838 www.tecodiag.com Email:tecodiag tecodiag.com
Standard/ Guidance Document Referenced (if applicable)
NCCLS EP5-A--Evaluation of Precision Performance of Clinical Chemistry Devices NCCLS EP6-A--- Evaluation of Linearity of Quantitative Analytical Methods NCCLS EP9-A---Method Comparison and Bias Estimation Using Patient Samples
BUN (Urea Nitrogen) is hydrolyzed by urease to produce ammonium and bicarbonate which resulted in the increase of solution conductivity which is directly proportional to the concentration of BUN present in the test, Calcium reacts with Arsenazo to form an bluish-purple colored chromophore which is measured spectrophotometrically at 650 nm. A reagent blank reading is taken just prior to sample injection, and a final absorbance reading is taken 21 seconds later. The differential absorbance, corrected for the reagent blank, is directly proportional to the calcium concentration. Creatinine from the sample combines with the reagent to produce a red color complex. Absorbance reading is taken at both 520nm and 560nm at 25.6 seconds after picking up sample. The differential absorbance has been shown to be a direct measure of the concentration of creatinine in the sample. SYSNCHRON CX3 determines glucose concentration by an oxygen rate method using a glucose Oxygen electrode. The rate of oxygen depletion is proportional to the glucose concentration in the samples.
1268 N. Lakeview Ave., Anaheim, CA 92807, U.S.A. Phone: (714) 693-7788 Toll Free: 1-800-222-9880 Fax: (714) 693-3838 www.tecodiag.com Email:tecodiaggtecodiag.com
The Teco CX3 Reagent Set for SYNCHRON CX System is substantially equivalent to other devices legally marketed in the United States. We have compared TECO Diagnostics Teco CX3 Reagent Set for SYNCHRON CX System to TECO Diagnostics Teco BUN Liquid, Teco Calcium Color, Teco Creatinine and Teco Glucose Liquid Reagent for Hitachi.The 510 (K) numbers are K981106, K864741, K880629, and K863926. Both devices are for the quantitative determination of the same analyst in serum.
Summary of Design Control Activities
The risk analysis method used to assess the impact of modifications was a Failure Modes and Effects Analysis (FMEA). The design verification tests that were performed as a result of this risk analysis assessment are listed in table l&2 below. The tests data can also be found in appendix C through F.
1. Table 1 - Performance Characteristics (Human Serum) BUN:
Feature Precision Within-Day Predicate Device Sample 1 N=20 Mean=12.9 SD=0.33 CV%=2.62 Sample I N=20 Mean=12.4 SD=0.15 CV%=1.2 Sample 2 N=20 Mean-51.8 SD=0.74 CV%=1.43 Sample 2 N=20 Mean=44.6 SD=0.75 CV%=1.67 Sample i N=25 Mean=l8 SD=0.46 CV%=2.5 Sample 1 N=25 Mean=18 SD=0.37 CV%-2.0 Candidate Device Sample 2 N-25 Mean=49 SD=0.91 CV%=1.9 Sample 2 N=25 Mean-49 SD=1.04 CV%=2.0 Sample 3 N=25 Mean=97.5 SD=1.3 CV%=1.3 Sample 3 N=25 Mean-98 SD=1.14 CV%-I.1
Day to Day
Teco CX3 Reagent Set, the acceptability of CV% in BUN test is within 8%
1268 N. Lakeview Ave., Anaheim, CA 92807, U.S.A. Phone: (714) 693-7788 Toll Free: 1-800-222-9880 Fax: (714) 693-3838 www.tecodiag.com EmaiI:tecodiag tecodiag.com
Sample 1 N=20 Mean=9.1 SD=0.39 CV%=4.3 Sample 1 N=20 Mean-9.2 SD=0.21 CV%=2.2 Sample 2 N=20 Mean=13.7 SD=0.02 CV%=0.2 Sample 2 N=20 Mean=13.3 SD=0.32 CV%=2.4
Sample 1 N-25 Mean=9.1 SD-0.08 CV%=0.9 Sample 1 N=25 Mean-9.1 SD=0.09 CV%=I.O Sample 2 N=25 Mean=12.3 SD-O.11 CV%-0.9 Sample 2 N=25 Mean=12.4 SD=O.14 CV%=1.1 Sample 3 N=25 Mean=14 SD=O.16 CV%-0.8 Sample 3 N=25 Mean=14 SD=O.14 CV%-1.0
Teco CX3 Reagent Set, the acceptability of CV% in Calcium test is within 5%
Creatinine: Predicate Device Feature
Precision Within-Day Sample 1 N=20 Mean-1.9 SD=0.05 CV%=2.6 Sample 1 N=20 Mean=2.0 SD=0.2 CV%=10 Sample 2 N=20 Mean=8.2 SD=0.6 CV%=7.3 Sample 2 N=20 Mean=8.2 SD=0.4 CV%=4.6
Sample 1 N=25 Mean=1.3 SD-0.04 CV%=2.7 Sample 1 N=25 Mean=l.3 SD-0.03 CV%=2.2 Sample 2 N=25 Mean-6.4 SD-O.11 CV%=1.6 Sample 2 N=25 Mean=6.6 SD-0.24 CV%-3.7 Sample 3 N=25 Mean-20 SD-0.35 CV%=1.7 Sample 3 N-25 Mean=19.9 SD-0.33 CV%=1.6
Teco CX3 Reagent Set, the acceptability of CV% in Creatinine test is within 10%
1268 N. Lakeview Ave., Anaheim, CA 92807, U.S.A. Phone: (714) 693-7788 Toll Free: 1-800-222-9880 Fax: (714) 693-3838 www.tecodiag.com Em aii:tecodiag tecodiag.com
Sample 1 N=20 Mean=87 SD-4.2 CV%=4.8 Sample I N=20 Mean=85 SD=3.7 CV%=4.3 Sample 2 N=20 Mean-282 SD-5.4 CV%=l.9 Sample 2 N=20 Mean=287 SD=9.6 CV%-3.3
Sample 1 N=25 Mean=78 SD=2.0 CV%=2.5 Sample 1 N=25 Mean=79 SD=2.6 CV%=3.3 Sample 2 N=25 Mean=255 SD=3.4 CV%=1.3 Sample 2 N=25 Mean-262 SD=5.9 CV%=2.2 Sample 3 N-25 Mean=735 SD=9.8 CV%=1.5 Sample 3 N=25 Mean=734 SD=8.7 CV%=1.2
Teco CX3 Reagent Set, the acceptability of CV% in Glucose test is within 10%
Table 2-Performance Characteristics (Human Serum) BUN:
3 to 102 mg/dL R:0.99 Y-1.00 -0.911
up to 80 mg/dL R=0.99 Y=0.962 X - 0.721
8-23 mg/L 15-39 mg/dL Expect values/ II Reference range Teco CX3 Reagent Set BUN test, Acceptance Criteria of Accuracy: r > 0.95; Slope: 0.97-1.1
1.0 to 15 mg/dL R-0.96 Y=i.07X +0.60 8.4-10.2 mg/dL
up to 20 mg/dL R=0.97 Y=0.94 X+ 0.53 8.5-10.5 mg/dL
Expect values/ Reference range Teco CX3 Reagent Set Calcium test, Acceptance Criteria of Accuracy: r > 0.90; Slope: 0.90-1.1
0.2 to24g/dL R-0.99 Y=0.97 -0.11 0.6-1.3 mg/dL
Up to 25 mg/dL
R=0.99 Y=0.96 X+ 0.06 Male: 0.9-1.5 mg/dL values/ Expect Female: 0.7-1.37 mg/dL range Reference Teeo CX3 Reagent Set Creatinine test, Acceptance Criteria of Accuracy: r > 0.96; Slope: 0.90-1.1
30 to 750 mg/dL R=0.99 Y=0.96X+0.36 70-105 mg/dL
up to 500 mg/dL R=0.99 Y=1.02 X+3.1 70-106 mg/dL
Expect values/ Reference range Teco CX3 Reagent Set Glucose test, Acceptance Criteria of Accuracy: r > 0.90; Slope: 0.85-1.1
DEPARTMENT OF HEALTH & HUMAN SERVICES
Public Health Service
Food and Drug Administration MAR 2005
Ms. Jian Vaeches Teco Diagnostics 1268 N. Lakeview Ave Anaheim, CA 92807
2098 Gaither Road Rockville MD 20850
k060120 Trade/Device Name: Teco CX3 Reagent Set for SYNCHRON CX System Regulation Number: 21 CFR 862.1770 Regulation Name: Urea nitrogen test system Regulatory Class: Class II Product Code: CDQ, CIC, CGX, CGA Dated: March 16, 2006 Received: March 16, 2006
Dear Ms. Vaeches: We have reviewed your Section 510(k) premarket notification of intent to market the device referenced above and have determined the device is substantially equivalent (for the indications for use stated in the enclosure) to legally marketed predicate devices marketed in interstate commerce prior to May 28, 1976, the enactment date of the Medical Device Amendments, or to devices that have been reclassified in accordance with the provisions of the Federal Food, Drug, and Cosmetic Act (Act) that do not require approval of a premarket approval application (PMA). You may, therefore, market the device, subject to the general controls provisions of the Act. The general controls provisions of the Act include requirements for annual registration, listing of devices, good manufacturing practice, labeling, and prohibitions against misbranding and adulteration. If your device is classified (see above) into either class II (Special Controls) or class III (PMA), it may be subject to such additional controls. Existing major regulations affecting your device can be found in Title 21, Code of Federal Regulations (CFR), Parts 800 to 895. In addition, FDA may publish further announcements concerning your device in the Federal Register. Please be advised that FDA's issuance of a substantial equivalence determination does not mean that FDA has made a determination that your device complies with other requirements of the Act or any Federal statutes and regulations administered by other Federal agencies. You must comply with all the Act's requirements, including, but not limited to: registration and listing (21 CFR Part 807); labeling (21 CFR Parts 801 and 809); and good manufacturing practice requirements as set forth in the quality systems (QS) regulation (21 CFR Part 820).
Page 2 This letter will allow you to begin marketing your device as described in your Section 510(k) premarket notification. The FDA finding of substantial equivalence of your device to a legally marketed predicate device results in a classification for your device and thus, permits your device to proceed to the market. If you desire specific information about the application of labeling requirements to your device, or questions on the promotion and advertising of your-device, please contact the Office of In Vitro Diagnostic Device Evaluation and Safety at (240) 276-0484. Also, please note the regulation entitled, "Misbranding by reference to premarket notification" (21CFR Part 807.97). You may obtain other general information on your responsibilities under the Act from the Division of Small Manufacturers, International and Consumer Assistance at its toll-free number (800) 638-2041 or (301) 443-6597 or at its Internet address http://www.fda.gov/cdrh/industry/support/index.html.
Alberto Guti eez, Ph.D.'
Division of Chemistry and Toxicology Office of In Vitro Diagnostic Device Evaluation and Safety Center for Devices and Radiological Health Enclosure
Indications for Use
510(k) Number (if known): Device Name: k060120
Teco CX3 Reagent Set for SYNCHRON CX System
Indications For Use: Teco CX3 Reagent Set for SYNCHRON CX System is intended for the
quantitative determination of BUN, Calcium, Creatinine and Glucose in serum on Beckman CX3 System. BUN measurements are used in the diagnosis and treatment of certain renal and metabolic diseases. Calcium measurements are used in the diagnosis and treatment of parathyroid diseases, a variety of bone diseases, chronic renal and tetany. Creatinine measurements are used in the diagnosis and treatment of renal diseases, in monitoring renal dialysis, and a calculation basis for measuring other urine analytes. Glucose measurements are used in the diagnosis and treatment of carbohydrate metabolism disorders, including diabetes mellitus, neonatal hypoglycemia and idiopathic hypoglycemia, and pancreatic islet carcinoma. This reagent set is intended for in vitro diagnostic use only.
(Part 21 CFR 801.109)
(PLEASE DO NOT WRITE BELOW THIS LINE-CONTINUE ON ANOTHER PAGE IF NEEDED) Concurrence of CDRH, Office of In Vitro Diagnostic Devices (OIVD)
Page 1 of
Cct ofIn Vitro Diagnostic Device Lxuluation and Safety
PB100 Kenwood 600 360055 CDX-GT180 KDC-W6527 24 FF Digital B1445V RTS-HE20 B4350 42PD6600 VS 530C Sview08 C24-E Faxphone 1600 KX-T123210BE GN 6110 F-V410 S-550 Yamaha FB01 Fantom-X7 DC C420 VP-D20 6230I 29K70 KDC-7080RV PL4260N 171MP Review MF 3560 UKF8001AXX DX7440 CD2302S Vivicam 6300 BTG 11 FAX 2316 IP 430 Controller SB4101 CL-21N11MJ Takara GP30 Deskjet 5743 HP 510 ECM-PC50 Kaleo Aculaser CX28 Systeme 5100 Behringer C-4 ME-33 LFV1060 KDL-22EX302 WAS32469GB DC280 PDR-609 HDR-SR10D CC290TW VX550 Design HD-Z1 Quadrasynth Plus Alesis D4 Commerciale 11 Cinema-U3000hybrid Coolpix 5700 Piranha MAX Geometry II Ninja 650R VGN-B1XP 2550DN HK690I SLV-SX70B LS-F1260HS 4200F ZGG646ITN Phaser 6110 AV-21JT5eu-av-21jt5 DCR-HC26E CP750 TTR50-2008 LRM RC-972 VGN-A115M Navisworks 2010 Altima-2004 E1245 3175 WX KDL-40NX700 ST6012 Party NWZ-S636F SHR-3040 VL-MS1 TG 100 FR-N3X Zoom 508 Mandarina Duck FS-820 A-8057 Fantom-X 6 KR-A3070 DVD-CM500
manuel d'instructions, Guide de l'utilisateur | Manual de instrucciones, Instrucciones de uso | Bedienungsanleitung, Bedienungsanleitung | Manual de Instruções, guia do usuário | инструкция | návod na použitie, Užívateľská príručka, návod k použití | bruksanvisningen | instrukcja, podręcznik użytkownika | kullanım kılavuzu, Kullanım | kézikönyv, használati útmutató | manuale di istruzioni, istruzioni d'uso | handleiding, gebruikershandleiding
1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30 31 32 33 34 35 36 37 38 39 40 41 42 43 44 45 46 47 48 49 50 51 52 53 54 55 56 57 58 59 60 61 62 63 64 65 66 67 68 69 70 71 72 73 74 75 76 77 78 79 80 81 82 83 84 85 86 87 88 89 90 91 92 93 94 95 96 97 98 99 100 101