Here you can find all about Citizen CH-432B like monitor and other informations. For example: .
Citizen CH-432B manual (user guide) is ready to download for free.
On the bottom of page users can write a review. If you own a Citizen CH-432B please write about it to help other people. [ Report abuse or wrong photo | Share your Citizen CH-432B photo ]
Citizen CH-432B, size: 117 KB
User reviews and opinions
No opinions have been provided. Be the first and add a new opinion/review.
DIGITAL BLOOD PRESSURE MONITOR
DI AS SY ST OL IC m m Hg TO LI
m Hg Cm
STA POW ER
- Eng 1 -
CLEANING METHOD OF CUFF
After cleaning the cuff with a neutral detergent, lightly rinse it with water and let it dry in the air. Hard rubbing or washing in a washing machine will cause air leakage. Also, be careful not to get water into the air hose.
General Remarks on blood pressure measurement
If the cuff is to be deflated quickly, depress POWER switch. The attached cuff is suited for an arm circumstance of 22cm to 32cm. When attaching the cuff, rubber cuff hose should be placed on the inside of your arm on top of the arterial. Self measurement is no therapy! In no case may you change the dosage of medication prescribed by your physician. In operation for blood pressure measurement, you should urinate and should remain relatively still for 10 to 15 minutes prior to measurement. Exercise, eating and drinking, smoking, etc., prior to measurement may affect the results. The blood pressure varies constantly throughout the day. Measurement should be made regularly at the same time each day. Do NOT be impressed by the results of one measurement. Keep a record of blood pressure variations. Many readings tell a story. Emotional stress may tend to cause pressure to rise. This Product should not be used to monitor cardiac arrhythmia.
Diastolic Blood Pressure
mmHg HYPERTENSION 89 HIGH NORMAL 85 NORMAL mmHg
Systolic Blood Pressure
For proper measurement of blood pressure
Take a deep breath five or six time and then relax before measuring blood pressure. If you are tense when taking the measurement, you will not get a valid reading. Blood pressure will be elevated when you are in a worrisome or irritated state, suffering from lack of sleep or constipation, or after exercise or a meal.
- Eng 2 -
Putting on the cuff is one of the most important tasks for ensuring proper measurement of blood pressure. Wind the cuff around the arm and tighten it so that one finger may be inserted between the cuff and arm. Do NOT measure blood pressure after bathing or drinking. When you feel like relieving yourself, you should do so before measuring blood pressure. Measure blood pressure where the room temperature is about 20C. Do NOT measure blood pressure when it is too cold (below 10C) or too hot (above 40C) in the room.
Do NOT measure blood pressure immediately after drinking coffee or tea or smoking. Measure blood pressure with a relaxed and quiet posture. Keep the center of the cuff at the level of your heart, and do NOT move your arm or talk during measurement. Do NOT keep on measuring so many times, as your arm will have built up congestion of blood, and you will not get a valid reading.
English Espaol Portugus
PRECAUTION FOR USE AND MAINTENANCE
Precautions for use
If you suffer from heart disease, high blood pressure or other circulatory disease, please consult your doctor first. If you feel abnormal pressure of the cuff during use or any other irregularity, reduce the pressure by immediately switching off the power or removing the batteries and consult the sales outlet where purchased. If you think the measured value is abnormal or if measuring makes you feel unwell, discontinue use and consult your doctor. Measurement of blood pressure may not be possible for someone with a weak pulse or arrhythmia. Congestion, swelling, etc. may occur in some people. If blood pressure is measured repeatedly, an accurate result will not be achieved. An interval of about 10 minutes should be allowed. The memory is erased if the batteries are removed for replacement, etc. and a record should be kept of any necessary data.
Precautions for maintenance
Do NOT store the blood pressure monitor where it will be exposed to direct sunlight, high temperature (over 60C), low temperature (below 20C), high relative humidity (over 95%) or in a dusty place. Do NOT expose the blood pressure monitor to under force or vibration, or drop it. Remove the battery/batteries when not in use for a long period. Do NOT disassemble the apparatus. Do NOT bend the cuff or air hose unnecessarily. NEVER wipe the blood pressure monitor with thinner or benzene, as they may damage it.
- Eng 3 -
When the monitor is very dirty, wipe it clean with a cloth moistened with Sterilizing alcohol or a neutral detergent, and then wipe it with a dry cloth. Do NOT use thinner or benzene.
Reference European standard:
This products complies with EC instructions 93/42/EEC (Medical Device Directive) and with the following standards: EN 1060-1: 1995 Noninvasive Sphygmomanometers, Part 1 General Requirements and EN 1060-3: 1997 Noninvasive Sphygmo-manometers, Part 3 Supplementary Requirements for Electromechanical Blood-Pressure Measuring Systems.
Nature and frequency of maintenance:
This products is designed for use over an extended period of time; however, it is generally recommended that it be inspected every two years to ensure proper function and performance.
Name: Address: Premier Precision Ltd. Block E, G-H, 6th floor, Phase 2, Yip Fat Factory Building, 73-75 Hoi Yuen Road, Kwun Tong, Kowloon, Hong Kong
Protect the nature environment:
Please help protect natural environment by respecting national and/or local recycling regulations when disposing of the batteries and the products at the end of their useful lives.
Name: Address: WvW elektronische Gerte Vertrieb GmbH. Robert-Koch-Strae 20, D-80538 Mnchen, GERMANY
- Eng 4 -
IDENTIFICATION OF PARTS
Tube socket Systolic blood pressure display section Diastolic blood pressure/ Pulse display section
DI AS SY IC STOL mm Hg
in E /m PULS
Cuff tube Air connector
- Eng 5 -
1. Attaching of Arm Cuff and Posture During Measurement
Wind the arm cuff around the left upper arm tightly so that bottom edge of the cuff is positioned about 1 2cm above the elbow joint. When you will wind the cuff, please wind the cuff next to the skin of possible. (If measurement is made by rolling the shirts or jacket up over the arm, this will constrict the upper arm and prevent accurate measurement.) Cuff tube should be placed on the inside of your arm as figure.
Unfasten and open the cuff as shown in the figure.
Please make sure that the cuff is fastened tightly. Note that uneven cuff winding will prevent proper measurements.
- Eng 6 -
2. Carrying out the Measurement
Rest the elbow on a table so that the section of the upper arm is at the height of the heart. Relax the lower arm across the table and remain still, since movement - artifacts may falsify the measuring result. Make measurements in either a seated or reclining posture with the palm turned upward.
Press the POWER / START switch.
SYSTOLIC mmHg DIASTOLIC mmHg
When the POWER / START switch is pressed, the lastmeasured systolic blood pressure and diastolic blood pressure are displayed for about 2 seconds.
Insert the air connector from the cuff into the tube socket at the mainframe until it is tightly connected.
At the first measurement after battery replacement the mark of Q is displayed for about 2 seconds.
- Eng 7 -
When the POWER / START switch pressing, the pressure level for measurement will change at optional level over 160 mmHg. But if the POWER / START switch is released before 160 mmHg, the pressure is heightened to 160mmHg.
When the pressure lowers by 15mmHg after pressurization stops, the mark of Q is displayed.
The pressure is automatically heightened to the level for measurement of the blood pressure based on the last-measured systolic blood pressure. At the first measurement after battery replacement the pressure is heightened to 160mmHg. If the level of the pressure is not enough, the pressure is automatically heightened to the level for measurement of the blood pressure again.
If pulse is recognized, the pressure is shifted to the diastolic blood pressure indicating section and change over, synchronized with the pulse.
- Eng 8 -
Automatic Pressure Setting
q This product automatically sets the pressure based on the previous measurement, so you can easily measure blood pressure by pressing one button without resetting the pressure for each measurement. If several people use the unit, we recommend using the manual settings as follows.
Manual Pressure Setting
When the measurement has been completed, the systolic blood pressure, the diastolic blood pressure and the pulse rate are alternately displayed and the air is discharged. The systolic blood pressure and the diastolic blood pressure are displayed for about 2 seconds and the pulse is displayed for about 1 second. The measured data are registered automatically, as long as they are not error. When a person with maximum blood pressure of 170 mmHg or higher uses the unit for the first time, or after changing batteries, measure blood pressure using the manual setting. If the automatic pressure setting is used, it may not work.
q When several people use the unit, set the pressure value manually. Please note that if the manual pressure setting is not used, pressure will be automatically applied based on the value set by the previous user. If the previous user had a high pressure setting, that high pressure will automatically be used for the current user. If you feel pain in your arm, press the POWER / START switch immediately to turn of the power. 1. Put the arm cuff around your arm, plug the air hose into the unit, and keep your arm at the level of your heart. 2. Keep the POWER / START switch depressed until the pressure display shows the optimal pressure setting. (The optimal pressure setting is about 40 mmHg higher than your usual maximum blood pressure.) 3. When the pressure display reaches the optimal pressure value, release the POWER / START switch. When the POWER / START switch is released pressurization will finish and measurement will start. Manual pressure settings can be selected between 160 mmHg - 280 mmHg. If you release the POWER / START switch before reaching 160 mmHg, the unit will pressurize to 160 mmHg. If the unit senses that the pressure is not high enough, it will apply more pressure.
Portugus Italiano Deutch Franais
- Eng 9 -
3. Turn Power OFF
Press the POWER / START switch to turn off.
Remove the battery cover at mainframe and replace four type AA batteries.
4. Subsequent Measurement
When a measurement is need again, 10 minutes, interval is necessary between measurements. Batteries are almost empty.
Batteries are normal.
5. Automatic Power OFF Function
If the unit is left without any operation after measurement, an Automatic Power-OFF Function runs to turn off the power after approximately 3 minutes. To continue measurement, press the POWER / START switch again. Note: The memory data are erased when you replace the batteries. The batteries that are included power the monitor. They may not last as long as indicated. If manganese batteries are used at a low room temperature, such as in the winter or for other reasons, the batteries may not function properly and you may not be able to measure your blood pressure. Please dispose of used batteries properly to protect the environmental.
6. Battery Replacement
If the batteries are almost empty, the mark will be displayed. In this case, batteries should be replaced with new one.
- Eng 10 -
DESCRIPTION OF DISPLAY MARKS
CONDITION / CAUSE Displayed when the blood pressure could not be measured accurately. (Some noises were detected by the movement of arm during measurement.)
Take off the cuff and wind it again, and carry out measurement once more. Do not move the arm during measurement.
Displayed when the pulse rate could not be measured.
Displayed when the batteries are almost empty.
Replace four batteries.
The unit has a trouble.
Take it to the store where it was purchased for testing.
- Eng 11 -
Model No. Measuring Principle Measurement Localization Cuff Arm Circumference Range Measurement Range Accuracy Pressure Pulse Pressure Pulse Pressure LCD Indication Pulse Marks LCD Switch Inflation Deflation Exhaust Rated Voltage Power Source Power Consumption Battery Life Automatic Power-Off Function Memory Function Dimension Weight Operating Environment Storage Environment Temperature Humidity Temperature Humidity Manganese Batteries Alkaline Batteries View Size Digit Size (Pressure) CH-432B Oscillometric Method Left Upper Arm Soft Cuff 22.0 to 32.0cm 0 to 280 mmHg 40 to 180 Pulse/min. 3 mmHg 5% of Reading Systolic blood pressure: 3 Digits Diastolic blood pressure: 3 Digits Alternately Pulse rate: 3 Digits Measuring Q Battery Empty 18(H) 61(W)mm 14(H) 6.5(W)mm POWER / START switch Automatic Inflation by Internal Pump Automatic Air-release Valve Electromagnetic Valve 6 VDC 4 AA Size Batteries 4W (Measuring) Approx. 200 times (160 mmHg, 1 time/day, 22C) Approx. 700 times Approx. 3 min. 1 Memory 129(W) 121.5(D) 55(T) [mm] Approx. 300g 10 to 40C 30 to 85%RH 20 to 60C 10 to 95%RH
* Specifications and designs may be changed without notice in the event of improvements being made.
- Eng 12 -
qCITIZEN is a registered trademark of Citizen Watch Co., Japan.
Consumer Electronics Division
CBM Bldg. 5F, 5-68-10, Nakano, Nakano-ku, Tokyo 164-0001, Japan Phone: (03)5345-7430 Fax: (03)5345-7431 E-mail: firstname.lastname@example.org
dabl Educational Trust
Device Manufacturer Category Protocol Reference Citizen UpperArm(Self) ESH
Device CH432B Mode Oscillometric
CotteUV,FaltenbacherVH,vonWillichW,BognerJR.Trialofvalidationoftwo devices for selfmeasurement of blood pressure according to the European Society of Hypertension International Protocol: the Citizen CH432B and the CitizenCH656CBloodPressureMonitoring2008;13:5562
Recruitment Screened Notstated Included RangesCorrect SexesCorrect 33 Yes Yes
ExcludedExplained None AgesCorrect DescriptionofMethods Procedure Description Results Description Poor Acceptable Yes Acceptable
An explanation, or even acknowledgement, for the dropinDBPpressuresismissing.
TheSBPplothas30,27&20visiblepoints(total78) in low medium and high ranges resp. The DBP plot has29,44&3visiblepoints(total76)inlowmedium and high ranges resp. Allowing for superimpositions thisislikelytobe38,57and4resp.Noexplanationis provided for the substantial drop in pressures from therecruitmentpressureforDBP.Expectedvaluesare 33 for each zone and the differences become significantlydifferentat22.Avalueof3or4cannot gounexplained.
2007 dablEducational Trust., 34 Main Street, Blackrock, Co. Dublin
Signoff Notes If a sentence acknowledging the shift in pressures from the recruitment pressures is supplied together with a table showing the range distribution for BP1,BP3,BP5andBP7thenthedevicemayberecommended. Date 20/05/2008
Series XPR IFP-195 Edition CT-W401R AP5VM-OV DUO-stereo SDR-S15 FAX-phone 1600 HW-C560S B4350 CPU-01015B XC-IS21T Vortex 14781 W VPC-E60EX Digidelay CDX-16 All-IN-ONE -TI-84 Plus Rapide LE40C652 Patchmate FY2300 15VL56 TX-51P15H 581 ID CDX-GT23W D-FJ041 SDM Civilization III Fcd2np Explorist 500 CQ-C1110W TXP50VT20E Temporis 26 Latitude X200 108 178 Jabra 9470 Eheim 3581 DIR-625 F45FD SU-600 ELP-3500 ACS54 TY-SP65p7 UB1204FX-PRO NN-GD379sepg 660HW-D SCP-55 RP100A Epson R-D1 SA-7500 VS900 KX-TG1311TR Modo 501 Alcatel 4035 Ducati 695 KD-S821R Alcatel 4059 Hsdsp2255A Keypad GT4 0C K 702 Aspire 9120 IC-F21 ABM 4900 BBA 2784 RL29thcts Graficko 3D PMP1000 Linux Eu R DVR-118L Sbcru151-00H TAS4011 SF-565P AL-1551CS L1950SQ Classic Contax TVS3 61-361 System Bolens G174 Scenic XL L-308S TG-3468 BMW Z4 INA-N333r-space-software Amplifier HL-7050N Grandprix 2004 Monitor PMC-303L CMP-kvmswit20 Hatchback Supergoo IT5614 Astro S6000 MS-7345
manuel d'instructions, Guide de l'utilisateur | Manual de instrucciones, Instrucciones de uso | Bedienungsanleitung, Bedienungsanleitung | Manual de Instruções, guia do usuário | инструкция | návod na použitie, Užívateľská príručka, návod k použití | bruksanvisningen | instrukcja, podręcznik użytkownika | kullanım kılavuzu, Kullanım | kézikönyv, használati útmutató | manuale di istruzioni, istruzioni d'uso | handleiding, gebruikershandleiding
1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30 31 32 33 34 35 36 37 38 39 40 41 42 43 44 45 46 47 48 49 50 51 52 53 54 55 56 57 58 59 60 61 62 63 64 65 66 67 68 69 70 71 72 73 74 75 76 77 78 79 80 81 82 83 84 85 86 87 88 89 90 91 92 93 94 95 96 97 98 99 100 101